Presentation on theme: "Journal Club Jan 2012. Hollow-fibre – coupling Different modes propagate in the fibre EH 11 EH 12 EH 13 EH 14 Nisoli, M. et al; Selected Topics in Quantum."— Presentation transcript:
Hollow-fibre – coupling Different modes propagate in the fibre EH 11 EH 12 EH 13 EH 14 Nisoli, M. et al; Selected Topics in Quantum Electronics, IEEE Journal of. 1998, 4, 414-420.
Hollow-fibre – mode discrimination / attenuation Different modesDifferent fibre radii small fibre radius => high intensity but low transmission long fibre => good mode discrimination but lower transmission
Hollow fibre – design considerations i.P limited by self-focusing ii.ionizaton limits minimal core radius iii.length of fiber is limited by propagation losses Vozzi, C. et al., 2005. Applied Physics B: Lasers and Optics, 80(3), pp.285-289 α ~ 0.45, β ~ 0.51 Hea min (Ar) = 1.6 a min (Ne) = 1.8 a min (He)
conventional Hollow fibre parameters L = 1 m ID = 250 µm Input: ~1 mJ with τ ~30 fs best compromise between transmission losses and broadening. L is limited by production and table. Transmission sensitive to radius (more bendy = less transmission) Nagy, T., Forster, M. & Simon, P., 2008 Appl. Opt., 47(18), pp.3264-3268. in principle: larger inner diameter and using a longer fiber to compensate for smaller intensity allows for better overall transmission
Flexible fiber – why? Nagy, T., Forster, M. & Simon, P., 2008 Appl. Opt., 47(18), pp.3264-3268. bending problem is eliminated by stretching the fibre on the ends.
Flexible fibers – what is this? ID/OD = 250µm / 360µm ID/OD = 320µm / 440µm
Flexible hollow fibers – setup HF: hollow fiber GT: glass tube VC: vacuum-tight cement OR: O-ring CR: clamping ring VT: vacuum tubing “After curing, the fiber is cut within the glued regions by a diamond tool.”
Flexible fibers – Results input: 1.1mJ, 70fs, 780nm 42% 1m 250µm Ar 300 mbar static fill F=9.2 64% 3m 320µm Ar 200 mbar static fill F = 9.7 F = Δω out /Δω in evacuated!! (=useless) close to evacuated a lot less then with no gas spot too small for 300 mbar of Argon? ionization at entrance? shifting of z-position? Obviously you put more effort into result you want to see Can you get better transmission with 1m?
Flexible fibers – Results 36% 1m 250µm Ar 500 mbar static fill “The ultimate spectral broadening with still regular spectral shape and acceptable spatial homogeneity” 48% 3m 320µm Ar 500 mbar static fill
Discussion Pros: easy to keep the fiber straight fibers don’t break that easily good transmission good beam profile after fiber Cons: increasing ID => longer tubes table size limited tiny fiber might be immediatly burnt at higher energies larger ID impractical for us?
Your consent to our cookies if you continue to use this website.