We think you have liked this presentation. If you wish to download it, please recommend it to your friends in any social system. Share buttons are a little bit lower. Thank you!
Presentation is loading. Please wait.
Published byKylie Lime
Modified over 2 years ago
Target Test Diagnostics Richard Brownsword Engineering, Rutherford Appleton Laboratory
Target Surface Velocities Problem : to measure surface displacement remotely in a time-resolved manner for an arbitrary surface material
Velocity Interferometry (VISAR) : laser interferometric technique time resolution may be ~ 1 ns compensates for changes in surface reflectivity
Velocity Interferometry (VISAR) : Laser Frequency ω Sample Velocity u(t) Fixed mirror Beamsplitter Etalon Length L Refractive index n Detector Fixed mirror
Velocity Interferometry (VISAR) : recombines beams at beamsplitter with delay = (2 L / c)( n - 1/n ) beat frequency (t) = ω [u(t) – u(t - )] / c time and velocity resolution determined by etalon (L, n) optical fibre etalons offer flexibility
Derivation of velocity : VISARs measure acceleration : Δu in time fringe constant K f = o / (ms -1 ) fringe count f c Δu u (t) = u (t - ) + Δu (t)
VISAR Simulations : surface velocity G. Skoro
VISAR Simulations : VISAR signal Detection limit
Background emission Expect source temperature ~ 2000 K VISAR laser = 1.55 m (1) Narrow-band interference filter ~ 500:1 discrimination (2) Background subtraction module
Conclusion Velocity interferometry : has potential for remote measurement of surface displacement for a range of different materials and surface velocities 0.01 ms -1 @ 10 ns ?
Frequency Scanning Interferometry (FSI) measurements
Gravitational Wave Astronomy Dr. Giles Hammond Institute for Gravitational Research SUPA, University of Glasgow Universität Jena, August 2010.
Frequency Scanned Interferometer Demonstration System Tim Blass, Jason Deibel, Sven Nyberg, Keith Riles, Haijun Yang University of Michigan, Ann Arbor.
Development of Experiment for Materials Studies at the Z-Beamlet Laser Facility Lasers enable us to study the state of materials and the dynamics of material.
Millimeter Wave Sensor: An Overview
Basic electromagnetics and interference Optics, Eugene Hecht, Chpt 3.
PERFORMANCE OF THE DELPHI REFRACTOMETER IN MONITORING THE RICH RADIATORS A. Filippas 1, E. Fokitis 1, S. Maltezos 1, K. Patrinos 1, and M. Davenport 2.
§ 4 Optical Fiber Sensors
University of Baghdad college of engineering electronics & comm.dept. Optical fiber as pressure senor Student name: Zaianb Raad.
Photo Acoustic Effect And its usage for spectroscopy.
教育部顧問室光通訊系統教育改進計畫台科大 師大 淡江 東南 萬能 教育部顧問室光通訊系統教育改進計畫 台科大 師大 淡江 東南 萬能 11 波長 (Wavelength) 的精準量測 The amplitude versus wavelength, can be measured by wavelength.
Fiber-Optic Accelerometer Using Wavefront-Splitting Interferometry Hsien-Chi Yeh & Shulian Zhang July 14, 2006.
3. Optical Coherence Tomography (OCT)
Thermo-elastic properties characterization by photothermal microscopy J.Jumel,F.Taillade and F.Lepoutre Eur. Phys. J. AP 23, Journal Club Presentation.
LiCAS Project: FSI Overview Richard Bingham, Edward Botcherby, Paul Coe, John Green, Grzegorz Grzelak, Ankush Mitra, John Nixon, Armin Reichold University.
Refractive index Petr Šafařík. Refractive index Petr Šafařík.
LumiCal Alignment System Status report Leszek Zawiejski, Tomasz Wojtoń, Arkadiusz Moszczyński Institute of Nuclear Physics PAN, Cracow 25th FCAL Collaboration.
U N C L A S S I F I E D Experiments and Simulations of Ablatively Driven Shock Waves in Gadolinium Richard Kraus, Eric Loomis, Shengnian Luo, Dennis Paisley,
Optical Coherence Tomography Zhongping Chen, Ph.D. Optical imaging in turbid media Coherence and interferometry Optical coherence.
UCLA Frequency-domain interferometry diagnostic system for the detection of relativistic plasma waves Catalin V. Filip, Electrical Engineering Department,
Beyond The Standard Quantum Limit B. W. Barr Institute for Gravitational Research University of Glasgow.
광섬유 센서를 이용한 변형 측정 김대현 Department of Aerospace Engineering Smart Structures and Composites Laboratory Optical Fiber Jacket ( = 300.
FT-IR stands for Fourier Transform Infrared, the preferred method of infrared spectroscopy. In infrared spectroscopy, IR radiation is passed through.
Diode laser The diode or semiconductor laser Diode lasers are extremely compact (0.3×0.2×0.1 mm), high efficiency (up to 40%), tuneable lasers, which have.
Piezo and retroflector calibration Understanding the details of interferometer performance.
Aaron Burg Azeem Meruani Michael Wickmann Robert Sandheinrich
Center for Radiative Shock Hydrodynamics Fall 2011 Review Experimental data from CRASH experiments Carolyn Kuranz.
Koji Arai – LIGO Laboratory / Caltech LIGO-G v2.
Interference of Light Ø It is generally He-Ne type that generates stable coherent light beam of two frequencies. one polarized vertically and another.
Lock-in amplifiers Signals and noise Frequency dependence of noise Low frequency ~ 1 / f –example: temperature (0.1 Hz), pressure.
Detection of weak optical signals D.R. Selviah, R.C. Coutinho, H.A. French and H.D. Griffiths Department of Electronic and Electrical Engineering, University.
IPP - Garching Reflectometry Diagnostics and Rational Surface Localization with Fast Swept Systems José Vicente
B. Gentry/GSFCGTWS 2/26/01 Doppler Wind Lidar Measurement Principles Bruce Gentry NASA / Goddard Space Flight Center based on a presentation made to the.
Interference and Diffraction
An Optical Receiver for Interplanetary Communications Jeremy Bailey.
CHRIS HOVDE, STEVE M. MASSICK and DAVID S. BOMSE
Tungsten wire & VISAR Goran Skoro 24 October 2008.
ENGINEERING APPLICATIONS Telecommunication Long Distance Communication Handle large number of channels Signalling purposes Computer Networks Connecting.
Sapphire for the LCGT project Eiichi Hirose ICRR, University of Tokyo Kyohei Watanabe, Norikatsu Mio PSC, University of Tokyo GT Advanced Technology, Sep.
MONALISA: Interferometric Position Monitor at the Nanometre Scale David Urner Paul Coe Matthew Warden Armin Reichold Oxford University.
T. Suehara, H. Yoda, T. Sanuki, Univ. of Tokyo, T. Kume, Y. Honda, T. Tauchi, High Energy Accelerator Research Organization (KEK) ATF2-IN2P3-KEK kick-off.
Ultrafast Experiments Hangwen Guo Solid State II Department of Physics & Astronomy, The University of Tennessee.
Electro-Optical Systems The almost last lecture of Weapons Class.
Imaging Science Fundamentals Chester F. Carlson Center for Imaging Science Interference and Holograms Introduction to 3D Images.
Brian Siller, Andrew Mills, Michael Porambo & Benjamin McCall Chemistry Department, University of Illinois at Urbana-Champaign.
Analysis of Phase Noise in a fiber-optic link
1© Manhattan Press (H.K.) Ltd. Reflection Refraction Refraction 12.1 Reflection and refraction Total internal reflection Total internal reflection.
An Approach to Stabilizing Large Telescopes for Stellar Interferometry It shakes like a…. [G. Vasisht, 31 March 2006] N. Di Lieto J. Sahlmann, G. Vasisht,
© 2017 SlidePlayer.com Inc. All rights reserved.