Presentation on theme: "Analysing X-ray data using GudrunX. Outline Planning an experiment – Absorption – Fluorescence – Beam size Data required Outline of analysis process Step."— Presentation transcript:
Analysing X-ray data using GudrunX
Outline Planning an experiment – Absorption – Fluorescence – Beam size Data required Outline of analysis process Step by step guide through analysis Practice with some data! – SiO 2 – H 2 O – Tellurite glass
Planning an experiment Before starting an experiment it is important to have a very good idea of your sample composition and density. This information will help identify any potential problems which may arise, such as: – Absorption +capillary size – Beam size – Measurements required – Fluorescence (We’ll return to this later) A good idea of potential problems will help you plan the length of your experiment too. – Strongly absorbing/weakly scattering or strongly fluorescent samples may require longer data collection Consider what your data will be used for and what quality you require.
Planning an experiment: Absorption If we accept 60% loss of flux, we can estimate the diameter of capillary to use: H 2 O : µ = ln(0.4)/ = 1.4 cm d ~ 1.8 cm Al 2 O 3 : µ = d ~ 2 mm GeO 2 : µ = d ~ 0.12 mm Y 2 O 5 : µ = d ~ 0.07 mm TeO 2 : µ = d ~ 0.18 mm PbO : µ = d ~ 0.03 mm Linear (µ) and mass (µ/ρ) absorption coefficients can be calculated from programs such as XOP (1) Area = πr 2 L =(πr 2 )/2r = π/4 d ~ 3/4 d 2r L
Increase Z → increased energy at which K edge occurs. for region around Ag (Z > Ag) µ/ρ < (Z
A Material chosen as a β filter must have an absorption edge which lies between the Kα and Kβ peaks. For an Ag tube, Rh is used. Planning an experiment: Absorption Example: β filter
Planning an experiment: Beam size Prog. Rec. slit Anti scatter slit X-ray tube K β filter Soller slit Prog. Div. slit Mask Soller slit Detector Capillary size PDSASS 0.51/81/4 1 1/2 1.51/ Table 1: PDS and ASS settings 240 mm (r) PDS angle (θ) Diameter of sample (L) PDS θ (rad) = L/r ASS= PDS x 2 PRS (mm) = L
Measurements needed Once the experimental setup has been decided up, three measurements are required – as with Neutron analysis, these are: Background Sample in capillary Empty capillary All these measurements need to be taken under the SAME CONDITIONS. The current set up is to collect data at 0.2° intervals from 3.2 – 156°. At each point, data is collected of 30 seconds. There is the option to collect two sets of data: Several repeat scans from 3.2 to 156° Additional scans from 35 to 156° to improve statistics at high Q
GudrunX: What does it do? Krogh-Moe – Norman normalisation Polarisation Absorption Compton scattering Calculating the coherent scattering Measured data background data
GudrunX: What does it do? sample 2 %diff SiO % Ga 2 O % Effect of normalisation: Calculating F(Q)
Installing GudrunX The X-ray diffractometer webpages can be found at OR as a link from the disordered materials group web page.
Instrument panel: The required files are all located in the gudrunX folder. User may wish to alter the Q range of the F(Q) produced, depending on the quality of the data. The Q max should be set to the final Q max you chose for you data.
Beam panel: Requires minimal alteration. Edit the beam size if the beam is smaller that the sample. Ensure the correct bremsstrahlung file is chosen.
Normalisation panel: User must choose which method of normalisation they wish to apply to the data. Altering Breit-Dirac factor and Overlap factor can give some improvements to the extracted F(Q). Maintain default values initially.
Sample background panel: Select an appropriate sample background panel. ‘Read data’ will display the information from the.XRDML file, including number of scans and the range of angles over which the chosen data set has been measured. Set sample background factor (between 0.9 and 1)
Sample specific information required: Once the instrument and background information has been checked, new tabs need to be added to give sample specific information. As with GUDRUN this includes a sample and a sample container tab. Information required includes: Sample specific information: Composition Effective density Sample size Fluorescence - a problem for elements in the same row as Ag (Rb – Te) Multiple scattering Experimental setup: Polarisation - 0 Compton scattering - 1 Bremsstrahlung - 0.4
Container panel: Composition, container size (inner and outer dimensions), effective density. For density either the measured effective density can be given, with a tweak factor = 0 Or the bulk density can be used with the tweak factor altered Effective density = bulk density/tweak factor
Sample panel: Basic information + fluorescence, multiple scattering etc. Ensure that packing fraction is sensible (measure or estimate it ~60%) Vary effective density and multiple scattering first, then bremsstrahlung. Only apply fluorescence for samples containing Rb – Te.
GudrunX: Output files.subcan X = 2 θ Y1 = experimental data Y2 = single atom scattering Y6 = Bremsstrahlung
GudrunX: Output files.soq X = Q Y1 = F(Q) F(Q) will have been normalised to either 2 or. Ensure that you have a record of which you used!
.gofr X = r Y1 = G(r) Quality of G(r) can be improved by varying parameters in GudrunX. Alternatively, the fourier transform software in Open Genie can be used. Daniel will be discussing the relationship between various correlation functions GudrunX: Output files
Fluorescence X-ray energy > absorption edge in sample → Fluorescence Fluorescence provides a background which is uniformly distributed across the angular range
Fluorescence Multiplying the data measured for the empty capillary and Ca/Sr glass data by a scale factor to match the Ca glass data (at high angle) gives: The shape of the capillary and calcium data are well matched. Problem with strontium sample.
Fluorescence However, if a constant background is subtracting from the Ca/Sr data and THEN the data is scaling: The characteristic X-ray shape is once again present in the data