Download presentation

Presentation is loading. Please wait.

Published bySamara Staver Modified over 4 years ago

1
RIGA Event Timer A032-ET Riga A032-ET in Graz

2
RIGA Event Timer A032-ET Eastbourne, October 2005 Graz uses routinely the Graz E.T. (Dassault Modules) 3.5 ps RMS / TI; < 2.5 ps non-linearity, 2.5 kHz max; Very good reliability; 5 years of operation now, Additionally, the new A032-ET from Riga now in Graz: 10 kHz Very useful in Graz for: Backup; additonal timing channel; Continuous control / verification of both ET‘s; In Progress: Measure Return Energy of each Return (via TI between comp./uncomp. C-SPAD output)

3
RIGA A032-ET vs. Graz E.T.: CAL Eastbourne, October 2005 Most important: Excellent Linearity of BOTH ET‘s !!! Deviations within 1-2 ps; In TI‘s from 100 ns (CAL:140 ns) to 150 ms (GPS) RMS Values: Riga: < 8 ps; Graz E.T.: < 3.5 ps; that means for the Complete SLR System in a Standard CALIBRATION: CAL RMS = 14 - 15 ps / 2.1 – 2.3 mm (with Graz E.T.) CAL RMS = 17 – 18 ps / 2.6 – 2.7 mm (with Riga ET) Both with standard setup, 2.2 Sigma Iteration etc...

4
RIGA A032-ET Test Results Non-Linearity: < 1 ps; fits perfectly also to Graz E.T.

5
RIGA A032-ET Test Results Warm-Up, 4.4 h Stability Test: < 1 ps drift after WA; Each dot: Mean of 900 Measurements

6
RIGA A032-ET Hardware

Similar presentations

Presentation is loading. Please wait....

OK

11 = This is the fact family. You say: 8+3=11 and 3+8=11

11 = This is the fact family. You say: 8+3=11 and 3+8=11

© 2018 SlidePlayer.com Inc.

All rights reserved.

To make this website work, we log user data and share it with processors. To use this website, you must agree to our Privacy Policy, including cookie policy.

Ads by Google