We think you have liked this presentation. If you wish to download it, please recommend it to your friends in any social system. Share buttons are a little bit lower. Thank you!
Presentation is loading. Please wait.
Published byJakob Ayles
Modified over 2 years ago
Page 1© Crown copyright 2004 Introduction to upper air measurements with radiosondes and other in situ observing systems John Nash, C. Gaffard,R. Smout and M. Smees Observation Development, Met Office, Exeter Integrated Ground-based Observing Systems Applications for Climate, Meteorology and Civil Protection 03-07 September 2007, L’Aquila, Italy
Page 2© Crown copyright 2004 Introduction Introduction Review of Radiosondes past and present Sampling provided by radiosondes Errors of radiosonde measurements [Aircraft measurements, references provided].
Page 3© Crown copyright 2004 Basic source on radiosondes and upper wind measurements Guide to Meteorological Instruments and Methods of Observation Seventh Edition [revised 2006] WMO- No8 Should be available in electronic form from WMO by end of 2007. Part I chapter 12 Measurement of upper air pressure, temperature and humidity chapter 13 Measurement of upper wind Part II chapter 10 Balloon Techniques
Page 4© Crown copyright 2004 1680 MHz radiotheodolite as used for tracking radiosondes in the USA for many years
Page 5© Crown copyright 2004 Various antenna for modern GPS radiosondes operating near 403 MHz Mauritius 2005.
Page 6© Crown copyright 2004 Upper wind Wind profiles from radiosondes are derived from tracking the displacement of the balloon from the launch site as a function of time. The radiosonde itself can be tracked using a radiotheodolite, secondary radar or reception of navigational signals by an antenna on the radiosonde. Terrestrial navigational signals Omega and Loran-C were commonly used from 1980 onwards, but now have mostly been replaced by GPS [Global Positioning System ] satellite signals. In some countries radars were used to track a target hung under the balloon The rate of ascent of the balloons can be between 5 and 8 ms -1, and the tracking techniques used with the older radiotheodolites allowed a resolution of about 2 minutes in wind near the ground and 4 minutes at best in the stratosphere. Now GPS signals allow a temporal resolution of a few seconds, but in fact winds are normally filtered longer than this to take out the rotation of the radiosondes in a pendulum under the balloon, since a suspension length of at least 20m is necessary to avoid contamination of the radiosonde measurements by the balloon at upper levels.
Page 7© Crown copyright 2004 Balloon usage The modern balloons routinely used by most meteorological services are either natural rubber latex or a synthetic latex.They expand as they ascend and will lift the radiosonde to heights between 20 and 26 km above the surface, before bursting. The weights of these balloons are usually between 200g and 500g For more expenditure on a larger balloon, weight around 1000g, heights in excess of 30 km [ pressures lower than 10 hPa ] can be obtained before burst. The lifting gas may be hydrogen or helium. Very few stations in the GCOS Upper Air network manage to observe regularly to a pressure of 5 hPa. Most modern radiosondes are not designed to observe accurately at pressures much lower than 5 hPa. However, observing heights as high as 40 km can be obtained using special plastic [non- expanding] balloons
Page 8© Crown copyright 2004 350g balloon +Vaisala RS80 GPS
Page 9© Crown copyright 2004 Review of progress with radiosondes and radiosondes in use In 1955, the radiosonde was the primary source of upper air information for meteorologists. Large numbers of people were employed at an upper air station. Four to six persons on a shift. All reports were produced by hand from traces on chart recorders or decoded from coded radiosonde signals by hand. Radio reception was relatively poor compared to modern systems, and this limited the ability to compare the output from different radiosondes flown together. Hence, the WMO Radiosonde Comparison held in Payerne in 1956 was unable to provide any serious insight into the relative performance of the radiosonde types, and the reference radiosondes tested in WMO Reference Radiosonde Comparison conducted in 1968 did not agree very well. In this period, the quality of the message was strongly dependent on the motivation of the staff to perform the calculations correctly.
Page 10© Crown copyright 2004 Assembling the radiosonde rig in 1955
Page 11© Crown copyright 2004 All ready to launch the radiosonde
Page 12© Crown copyright 2004 UK radiosonde used for several decades prior to 1978, Note windmill turned a mechanical switch which selected which sensor assembly controlled the transmission to the ground. Thus, sensor repeat sampling was slow and not at a constant rate.
Page 13© Crown copyright 2004 Pressure Temperature Relative humidity The sensors were mechanically linked to a variable inductance Which controlled the frequency of transmissions to the ground at 28 MHz.
Page 14© Crown copyright 2004 Indian Radiosonde groundstation in 1999
Page 15© Crown copyright 2004 Coding of messages Early radiosonde signals were transferred to special working diagrams after decoding or were made available on chart recorder outputs. The messsages sent were derived by hand. So the results were reported in a code that allowed the values of temperature and dewpoint depression and wind to be reported at a set of standard pressure levels, e.g. 850, 700, 500, 300, 200, 100, etc. hPa. Here the values were supposed to be those actually measured. In addition the message contained an additional set of levels that allowed the temperature profile structure to be reproduced to ± 1 C in the troposphere and ± 2 C in the stratosphere, and the relative humidity profile to ± 15 per cent. When radiosonde observations started these coding errors were probably not much different to the errors in the sensor measurements, but this is not the case today. The sensor sampling rates in the old radiosondes were not high, but now detailed profile observations with the sensors sampled every second, can be exploited by taking detailed data from the ground system archives and not relying on the approximations in the TEMP code. A new BUFR code representing the detailed data is currently in the process of implementation.
Page 16© Crown copyright 2004 By early 1980’s Information from satellite remote sensing was expected to replace radiosondes by 1990. However, advances in electronics were allowing the introduction of improved data transmission and reception, electronic multiplexing of sensor output, improved sensors and affordable computers to process data semi- automatically or automatically. Thus, those countries able to take advantage of these innovations were able to change the quality and accuracy of radiosonde output, and to make it very much less dependent on the operator motivation.
Page 17© Crown copyright 2004 One type of Russian radiosonde still in use
Page 18© Crown copyright 2004 Goldbeaters skin Humidity sensor Support wires of T sensor too thick No pressure sensor
Page 19© Crown copyright 2004 Modern type of 1680 MHz radiotheodolite Meteorit II, secondary radar, now obsolete
Page 20© Crown copyright 2004 Chinese secondary radar
Page 21© Crown copyright 2004 Coding drum Temperature sensor, bimetallic spiral Chinese radiosonde, phased out 2005 -2007 replaced by modern system with accuracy much higher
Page 22© Crown copyright 2004 First system where operators could rest during an ascent 1984, Beaufort Park, UK
Page 23© Crown copyright 2004 The size of radiosondes in 1984 during the first modern WMO test is much larger than those used now and the measurements less accurate
Page 24© Crown copyright 2004 In recent years A new generation of radiosondes has been introduced exploiting ASIC [Application Specific Integrated Circuits] developments. This allows the use of decoded GPS signals, giving very accurate wind measurements and accurate height measurements. Occupied bandwidth has been reduced on standard radiosondes, and transmissions are more stable than in earlier standard designs. Temperature and relative humidity sensors have also been improved to extend the useful vertical range of both measurements, so reliable temperatures can be measured to pressures as low as 5 hPa and reliable relative humidity to temperatures as low as -70 deg C. Aircraft measurements are expected to reduce the number of regular radiosonde launches Ground-based remote sensing systems are being developed to provide continuous upper –air measurements at high temporal resolution, but often lower vertical resolution.
Page 25© Crown copyright 2004 Two examples of modern GPS radiosondes Internal electronics, 2006
Page 26© Crown copyright 2004 Summary The design and characteristics of radiosondes in use has changed radically with time, so that representations of radiosonde systems stuck in the past is largely untrue. However in some countries, relatively poor systems still remain, so that not all of the 700 civilian stations making measurements daily have the same error characteristics The better radiosonde are able to meet the most exacting user requirements for all operational use and scientific studies, apart from possibly the very high accuracy requested for relative humidity in the upper troposphere for climate trend studies. Detailed description of the various radiosonde observing principles can be found in the CIMO Guide. A knowledge of radiosonde errors is essential when using radiosondes to compare with remote sensing measurements. This includes an understanding of the representativeness errors from the rapid ascent of the balloon through the vertical structure of the atmosphere, usually 1 km in the vertical in about 3 minutes.
Page 1© Crown copyright 2004 Introduction to upper air measurements with radiosondes and other in situ observing systems  John Nash, C. Gaffard,R. Smout.
1 00/XXXX © Crown copyright Introduction to performance of modern radiosondes based on WMO Radiosonde Comparison results, Temperature John Nash Upper Air.
Page 1© Crown copyright 2004 Results of the RS92 Acceptance Test performed by the Met Office (UK) Richard Smout, John Nash, Mark Smees, Darren Lyth, May.
1 Radiosondes 1.History of upper air measurements 2.Radiosondes (sensors, calibration, telemetry,multiplexing) 3.The Vaisala radiosondes 4.Special radiosondes.
Storey: Electrical & Electronic Systems © Pearson Education Limited 2004 OHT 7.1 Control and Feedback Introduction Open-loop and Closed-loop Systems Automatic.
Weather Forecasting This chapter discusses: 1.Various weather forecasting methods, their tools, and forecasting accuracy and skill 2.Images for the forecasting.
Federal Department of Home Affairs FDHA Federal Office of Meteorology and Climatology MeteoSwiss TECO-2006, WMO, Geneva, Global Criteria for.
Page 1© Crown copyright 2004 Introduction to upper air measurements with radiosondes and other in situ observing systems  Factors affecting comparisons.
Wind profilers and radio contamination issues Dominique Ruffieux MeteoSwiss Aerological Station of Payerne Wind profiler - how it works - examples Frequency.
© Crown copyright Met Office E-AMDAR evaluation. Mark Smees & Tim Oakley, Met Office, May 2008.
SIMS-201 The Telephone System Wired and Wireless.
We will resume in: 25 Minutes We will resume in: 24 Minutes.
Abstract has 6 upper air stations for GPS S SR2K2 Modemwith Radiosonde M2K2_DC, 6 Upper Air stations of RDF Radiotidolite RT20A ( Vaisala ) and three Upper.
Vaisala GPS station Egyptian Meteorological Authority (EMA) Koubry ELQuobba EMA, Cairo, Egypt, P. BOX : Fax :
Equal or Not
1 Review of WMO test results on the accuracy of radiosonde relative humidity sensors John Nash Observing Methods Technology Centre, Development, Met Office.
25 seconds left….. 24 seconds left….. 23 seconds left…..
Vaisala RS92 Radiosondes offer a high level of GPS performance with a reliable telemetry link Hannu Jauhiainen, Matti Lehmuskero, Jussi Åkerberg TECO 2005.
Sources of Weather Data How do we measure and predict the weather?
U.S. Radiosondes Jan. 2000, NWS awarded contracts to two radiosonde manufacturers, Sippican and InterMet Systems, for the development and submission of.
Radiosondes ATS May 5th Balloon Launch. What is a Radiosonde? A radiosonde is a balloon-based instrument platform with radio transmitting capabilities.
Instruments. In Situ In situ instruments measure what is occurring in their immediate proximity. E.g., a thermometer or a wind vane. Remote sensing uses.
© TAFE MECAT 2008 Chapter 6(b) Where & how we take measurements.
World Meteorological OrganizationIntergovernmental Oceanographic Commission of UNESCO Ship Observations Team ~ integrating and coordinating international.
Chap 7-1 Statistics for Business and Economics, 6e © 2007 Pearson Education, Inc. Chapter 7 Sampling and Sampling Distributions Statistics for Business.
Introduction to data assimilation in meteorology Pierre Brousseau, Ludovic Auger ATMO 08,Alghero, september 2008.
S-Curves & the Zero Bug Bounce: Plotting Your Way to More Effective Test Management Presented By: Shaun Bradshaw Director of Quality Solutions Questcon.
Part 5. Human Activities Chapter 13 Weather Forecasting and Analysis.
1 Volume measures and Rebasing of National Accounts Training Workshop on System of National Accounts for ECO Member Countries October 2012, Tehran,
50 th Anniversary of Operational Numerical Weather Prediction John Jones Deputy Assistant Administrator for Weather Services June 15, 2004 University of.
Section 12.4 – Weather Analysis and Prediction 8 th Grade Earth and Space Science Class Notes.
Jeopardy Topic 1Topic Q 1Q 6Q 11Q 16Q 21 Q 2Q 7Q 12Q 17Q 22 Q 3Q 8Q 13Q 18Q 23 Q 4Q 9Q 14Q 19Q 24 Q 5Q 10Q 15Q 20Q 25 Final Jeopardy.
1 Ground Based Meteorological Radars Presented By: David Franc NOAAs National Weather Service September 2005.
Report to WCRP Observations and Assimilation Panel David Goodrich Director, GCOS Secretariat Towards a GCOS Reference Upper Air Network.
PMO-III, March 2006, Hamburg, Germany. Other PMO Activities.
Liang, Introduction to Java Programming, Eighth Edition, (c) 2011 Pearson Education, Inc. All rights reserved Chapter 3 Loops.
1 An introduction to Ultraviolet/Visible Absorption Spectroscopy Lecture 24.
Chapter 13 How Cells Obtain Energy from Food Essential Cell Biology Third Edition Copyright © Garland Science 2010.
Slide 1The Wave Model (ECWAM) 2. The WAM Model: Solves energy balance equation, including S nonlin WAM Group (early 1980s) State of the art models could.
Low-tropospheric wind profilers and radio contamination issues Dominique Ruffieux MeteoSwiss Aerological Station of Payerne Wind profiler - how it works.
1 RA III - Regional Training Seminar on CLIMAT&CLIMAT TEMP Reporting Buenos Aires, Argentina, 25 – 27 October 2006 Status of observing programmes in RA.
1 Proposed new uses for the Ceilometer Network Christine Chiu Ewan OConner, Robin Hogan, James Holmes University of Reading.
1 RA I Sub-Regional Training Seminar on CLIMAT&CLIMAT TEMP Reporting Casablanca, Morocco, 20 – 22 December 2005 Status of observing programmes in RA I.
Meteorological Observatory Lindenberg – Richard Assmann Observatory The GCOS Reference Upper Air Network.
1 NWS-COMET Hydrometeorology Course 15 – 30 June 1999 Meteorology Primer.
Stratospheric Measurements: Microwave Sounders I. Current Methods – MSU4/AMSU9 Diurnal Adjustment Merging II. Problems and Limitations III. Other AMSU.
The Evolution and Development of the United States NOAA National Weather Service Universal Radiosonde Replacement System TECO 2005 World Meteorological.
FIGURE 11.1 Circuit for Example Curtis Johnson Process Control Instrumentation Technology, 8e] Copyright ©2006 by Pearson Education, Inc. Upper Saddle.
© 2017 SlidePlayer.com Inc. All rights reserved.