We think you have liked this presentation. If you wish to download it, please recommend it to your friends in any social system. Share buttons are a little bit lower. Thank you!
Presentation is loading. Please wait.
Published byAutumn Lloyd
Modified over 3 years ago
Sterile Closure Screw Step by Step
Sterile Closure Screw PackagingDescription: Sterile “four in one” package (v4) Each closure screw can be opened individually Availability: RC: in heights 0 and 0.5 mm NC: in heights 0 and 0.5 mm Education
Step 1 - Opening the outer packagingInstructions for use and four blisters in the outer packaging Outer packaging is not sterile A B C D Education
Step 2 - Detaching a blister packBend single blister pack back and forth a few times (B) Detach one of the blister packs (C,D) A B C D Education
Step 3 - Opening a blister packPeel off blister foil (A,B) Remove the press pad (C) Do not contaminate closure screw Place open blister on an unsterile base A B C D Education
Step 4 - Picking up sterile closure screwClosure screw is fixated upright in blister pack (A) Pick up closure screw with SCS screwdriver Friction fit to secure closure screw B A C D Education
Impression procedure. STRAUMANN 2 Education Product portfolio for bone level implant impression procedures Color coding per implant connection Ø 3.3 mm.
48 hour tool Step by Step. STRAUMANN 2 Education Purpose of 48-hour tool Available for NC and RC Removal of placed implant from patient Substitute.
After Receiving this error code, Follow these steps to correct the problem… Click for next slide.
Anatomic Abutment. STRAUMANN 2 Education Anatomic Abutment Concept Component materials Anatomic Abutment (Ti) Basal Screw (TAN)* Benefits Excellent.
Wash hands using standard precautions techniques..
AB 11 22 33 44 55 66 77 88 99 10 20 19 18 17 16 15 14 13 12 11 21 22 23 24 25 26 27 28.
The reading is 7.38 mm. The reading is 7.72 mm.
A Review of the Basics n OraSure ® device n Collection procedures.
L.C. Technology ROBOT ASSEMBLY. Base Assembly BASE File off raised tabs on each side of Servo. Place Servo in vice between Fibre Clamps. Raised Tabs Repeat.
$100 $200 $300 $400 $500 $100 $200 $300 $400 $500 $100 $200 $300 $400 $500 $100 $200 $300 $400 $500 $100 $200 $300 $400 $500 $100 $200 $300.
Filter Changing Procedures For TEOM Series 1400ab and Model 8500 FDMS (B & C Types)
Straumann ® Bone Level Implant Surgical Product Portfolio.
Presented by: Sana’a AL-Sulami Teacher assistant Nursing department.
PS2200B/3000B Battery Replacement Procedure. 1.Disconnect the wire connecting the UPS and the battery pack. 2.Pull off the battery pack front panel.
Gold Abutmentfor bridge. STRAUMANN 2 Education Component materials Gold Abutment (non-oxidizing Gold) Basal Screw (TAN)* Modeling Aid (Polymer)
Perpendicular Lines. What is to be learned? The rule connecting gradients and perpendicular lines.
Gold Abutment for crown. STRAUMANN 2 Education For NC and RC Gold abutment Product portfolio NCRC.
Replacing the Main Monitor 1.Open Lower Main Door, and turn off machine power 2.Open the upper main door 3.Disconnect the video and power plugs from the.
- I chose to do the Option 1 Internship, which had 260 hours of work, and was paid. - The internship went from an internship to a job. - I now am working.
Straumann ® Bone Level Implant Prosthetic Product Portfolio.
Reconstitution of Dryvax® Smallpox Vaccine As taken from the Package Insert, Dec
SSD/HDD Upgrade Kit SJ25CK3. All-in-one Upgrade Kit – SJ25CK3 Product SJ25CK3 Size 129.5mm x 80.8mm x 18.8mm (enclosure) Weight 108g Connector USB 3.0.
GHP 2014 Avery Andrews Archish Chaturvedi Nandan Dave.
STEP 1 Take one piece of the adhesive backing sheets and one Kenrico detox pad.
Mary Beth Lemoine Media 6467 – UWG Spring Have you read the instruction manual? Make sure that you have the correct bulb for your overhead projector.
Get Easy Battery Replacement Guide at Battfix.com
IOPtiMate™ OT-135P2 Technical Training – Part I System Description, Installation & Operation IOP (07)
Australian inhibitor / Safe pack ARN Date 21 Mar 2012 Howitzer M198 Field Gun -Long term Storage.
Recombinant DNA Bacterial Transformation Teacher Preparation Sterile Materials.
2G BBULBE PASS Instructions-5 Replacement of Heater.
Target 3D About of the implant library Target 3D in Exocad Software Version Target 3D.
Vocabulary. Motion Definition – the act of moving Examples – A swing moves back and forth A merry-go-round moves around and around.
10 Easy steps to building your computer By Chris Holzworth & Michael Barnett.
Easy Reach 7.0 Honda Element Packing List QtyDescriptionPacked 115ft 10ga Red Wire 16ft 10ga Black Wire 140amp Breaker 23/8 Yellow Ring Terminals 25/16.
Mold Removal New Jersey. SURFACE SAMPLING FOR MOLD Surface sampling is the most common technique used to test surfaces for mold during a mold inspection.
LSM780 Live Cell Turn On/Turn Off Instructions Kim Peifley 07/13/15.
Universal Holding Cabinet Two screws hold each side; four screws secure the top.
Coil-pack assembly procedure Paolo Ferracin LQS01 readiness review #3 Assembly LBNL July 28, 2009.
ZIPP 404 BEARING REPLACMENT GUIDE Tools Needed: 1.Bearing Press with bushing to fit 6803 bearing size. 2.Hammer 3.Flat head screwdriver 4.Bearing puller.
Savanah Raised Toilet Seat Update. Product History Original seat launched in 1995/96; 5cm and 10cm without lids were the only products Designed in the.
2G BBULBE PASS Instructions-4 Replacement of bubble generator.
Aseptic Technique: The sterile field, gloving and gowning September 2012 Innovating for life. UC EE.
Asepsis Gloving. When to Wear Sterile Gloves Sterile gloves should be worn when there is direct contact with other sterile supplies and equipment Sterile.
Example 4 Using Multiplication Properties SOLUTION Identity property of multiplication () 16 a. 6 = Find the product. () 16 a.b. 15– () 0 Multiplication.
27/9/2007, T Water =19°C No Box screwed onto cooling block: T min =37°C, T max =48 °C.
PS500B/800B Battery Replacement Procedure. 1.Remove the two screws at the bottom of the front panel. 2.Push down then remove the front panel.
Handy Store Fixtures Gondola Dolly Assembly Instructions.
How to Perform Preventive Maintenance of the Personal Computer Properly Shut Down of the Computer Properly Shut Down of the Computer Unplug all Connection.
LAB #2 CARBURETOR REMOVAL AND REPLACEMENT. Remove Cover Using a #10 star tip and a bit driver, remove the four screws to recoil cover.
© 2017 SlidePlayer.com Inc. All rights reserved.