We think you have liked this presentation. If you wish to download it, please recommend it to your friends in any social system. Share buttons are a little bit lower. Thank you!
Presentation is loading. Please wait.
Published byBruce Malson
Modified over 2 years ago
© 2011 Pearson Education, Inc. GAS CHROMATOGRAPHY CHAPTER 21
© 2011 Pearson Education, Inc. Gas Chromatography Produces Chromatograms GC is used to identify, (compare to standards) separate, and quantify (use peak area) samples.
© 2011 Pearson Education, Inc. Stable, but Volatile Analytes Volatility is related to vapor pressure and boiling point. Larger molecules (> 600g/mol) are not volatile enough for GC. Boiling points < 500 o C help, as does low polarity. Heated injection and separation may degrade some compounds. Thermal stability is required. The structure of the analyte could be changed, in a known way, to increase volatility Derivatization.
© 2011 Pearson Education, Inc. Factors That Determine Retention in GC Retention is related to the relative time the analyte is in the mobile phase compared to the stationary phase. Retention is affected by volatility, column temperature, and the degree of interaction with the stationary phase. More volatile analyte elutes first (volatility increases make retention decrease). Note the homolog alkane series.
© 2011 Pearson Education, Inc. Factors That Determine Retention in GC TEMPERTURE: Decreasing column temperature (via oven) leads to longer retention times due to a drop in volatility. Note this effect in the chromatograms where temperature is changed on the same system:
© 2011 Pearson Education, Inc. Factors That Determine Retention in GC Kovats Retention Index (I): retention of an analyte on one column is compared to the retention seen at the same temperature and column for a series of alkanes. t Rz (adjusted retention time for n-alkane that elutes just prior to analyte) t R(z+1) (adjusted retention time for n-alkane that elutes just after the analyte) z refers to the number of carbon atoms in the homolog series. Exercise 21.3 allows practice for Kovats index.
© 2011 Pearson Education, Inc. Factors That Determine Retention in GC Interactions with the stationary phase can affect retention time….Examine the polarity of analytes and the stationary phase for interactions. Which of these two is more polar? Which would tend to retain polar analytes?
© 2011 Pearson Education, Inc. Use Kovats to Compare Differences in Retention Time Related to Polarity
© 2011 Pearson Education, Inc. Mobile Phases and Elution Methods In the mobile phase, gas is low density and moves quickly through the column, giving rise to narrow peaks and quick separations. Low viscosity permits the use of longer columns (without undue pressure build up). Move solutes through with no or minimal interaction with analytes. (He, N 2, and Ar common) The carrier gas (mobile phase) must be high purity to avoid contaminating the sample and possibly breaking down column materials.
© 2011 Pearson Education, Inc. Mobile Phases and Elution Methods For simple samples a constant temperature may be maintained during the analysis (isothermal method). Complex samples often contain analytes with a wide range of volatility, causing a wide range of retention times. The figure shows, using the isothermal method, that some analytes emerge quickly, others much longer. Also, resolution varies.
© 2011 Pearson Education, Inc. Mobile Phases and Elution Methods Temperature programming offers a solution to thegeneral elution problem (separating with reasonable resolution in reasonable time). Start chromatography at a lower temperature, then gradually ramp up during analysisnote the four regions.
© 2011 Pearson Education, Inc. GC Supports and Stationary Phases Columns may be discussed/compared by noting if they are packed or open-tubular. Packed columns are filled with small, absorbent particles or coated particles. Use a glass or metal column ~ 1–2 meters in length and a few mm wide. Silica (SiO 2 ) is used to make diatomaceous earth, which can be coated for use in Gas-Liquid chromatography. Molecular sieves or porous polymers are used with gas-solid chromatography. Packed columns have typical applications in simplier systems. A large surface area allows larger sample injection but has lower efficiencies.
© 2011 Pearson Education, Inc. GC Supports and Stationary Phases An open-tubular column (capillary column) has a stationary phase coated on its interior. These are much longer than packed columns (10–100m) with small diameters (0.1–0.75 mm). They also tend to be faster, have better resolution, and better efficiency. WCOT SCOT PLOT
© 2011 Pearson Education, Inc. GC Supports and Stationary Phases Compare packed and open-tubular columns:
© 2011 Pearson Education, Inc. Take a Closer Look Diatomaceous earth produces high surface area packing material. Molecular sieve have (zeolite) well defined pore size/structure.
© 2011 Pearson Education, Inc. Gas-Liquid Chromatography Gas-solid chromatography utilizes solid, absorbent material, and molecular sieves with good applications to small hydrocarbons and gases. The surface area, the size of pores, and functional groups on the supports surface determine separation. GSC is not as common as GLC, but can be packed or open. Gas-Liquid Chromatography utilizes high viscosity and low volatility. Its easy to place the liquid spread over the support. The liquid coating is often based on polysiloxane with varying molar masses ( 1000–10 6 g/mol).
© 2011 Pearson Education, Inc. Bonded Phase Support Some column bleed occurs during heating when the liquid stationary phase breaks down or vaporizes. This affects retention and even detectors. If the liquid can be covalently bonded to the solid support, a bonded phase support is formed, which is more stable (allows higher temp to be used.) This is formed by reacting a polysiloxane liquid phase with a silanol group on a solid silica support. A cross-linked stationary phase can also be used.
© 2011 Pearson Education, Inc. Stationary Phases and Polarity
© 2011 Pearson Education, Inc. Detectors: Thermal Conductivity TCD can be used for organic and inorganic analytes. The key aspect is the ability of the carrier gas and the analytes to change the conductivity of a wire filament, which will vary with different analytes. The carrier gas should have different thermal conductivity of analytes. TCD is a non-destructive type of detection that uses a Wheatstone bridge style. Downsides are the response to impurities, leakage in air, and poor response to LOD.
© 2011 Pearson Education, Inc. Wheatstone Bridge TCD Most common carrier
© 2011 Pearson Education, Inc. Detectors: Flame Ionization FID uses fuel mixed with carrier and organic analyte. Analyte forms ions in the flame. Cations from the flame are gathered by the negative electrodeproduces a current. Advantage:inorganics do not respond (He carrier gas), so the low background signal allows for LOD 100- to 1000-fold lower then TCD. Disadvantage: destructive
© 2011 Pearson Education, Inc. Detectors: Selective Detectors Specific for specific types of chemicals (i.e., nitrogen- phosphorus detector, NPD) Measures ions produced from eluting N or P compounds, but generates e – from a heated surface (not flame) that combine with electronegative elements to form negative ions Good selectivity, good LOD, but must periodically change heated material
© 2011 Pearson Education, Inc. Detectors: Electron Capture Detects electronegative atoms or groups (Cl, -NO 2 ) and also polynuclear aromatics and conjugated carbonyl compounds These groups capture e – that are produced from nuclear radiation from 3 H Carrier gas hit by these e – can release secondary e –, which will be absorbed by analyte Has good LOD, but a narrow linear range and requires radioactive source
© 2011 Pearson Education, Inc. Detectors: Mass Spectrometry Detects and measures by converting eluting analytes into gas phase ions (forming a molecular ion or fragmenting analyte [and ionizing]. Compare patterns of ions and fragments to known values. Intensity relates to amounts. A portion of analytes is converted to anions via electron impact or chemical impact (softer, less fragments). Gas phase ions are separated by mass/charge ratio using a quadrupole mass analyzer. (Uses four parallel rods with well-defined potentials so that only certain mass/charge species may pass [sorted]) See Exercise 21.4.
© 2011 Pearson Education, Inc. Detectors: Mass Spectrometry A Mass spectrum plots intensity of mass/charge vs. timecould be set a certain retention time in a GC. Mass chromatogram
© 2011 Pearson Education, Inc. Sample Injection and Pretreatment Gas Sample: if present at moderate to high concentration, may be directly put into column via gas-tight syringe. Another technique is to use a gas-tight valve: For trace levels like volatile organic compounds (VOC), the sample may need to be preconcentrated by passing it through a solid-absorbing cartridge or using a cryogenic trap.
© 2011 Pearson Education, Inc. Sample Injection and Pretreatment Liquid Samples: direct injection of a relatively large sample (6-10 mL). A calibrated microsyringe may be injected through a gas-tight septum into a heated chamber that volatilizes the sample. Note that there are direct, split, and splitless column styles. Sometimes pretreatment requires dervitatization or moving the sample into a different solvent (remove from H 2 O). Static or Dynamic Headspace analysis of vapor phase above a sample avoids water in the sample.
© 2011 Pearson Education, Inc. Sample Injection and Pretreatment Solid Samples: First extract compounds of interest from a solid matrix by liquid-liquid extraction or supercritical fluid. Samples are placed in solvent, then treated as liquid samples. Thermal desorption may be used on some solids as analytes may be collected during heating of a solid. Pyrolysis GC is employed for substances that are not volatile and cannot be easily derivatized to volatile forms. The solid is heated in a controlled way to break it into smaller, more volatile pieces separated by the column to form a pyrogram, which can be matched to known standards.
Analytical Separations Gas Chromatography. Instrument.
Chapter 27 Gas Chromatography 1. Principles Two types -Gas-solid chromatography (limited because of semipermanent retention of polar molecules) - Gas-liquid.
1 Chapter 24 GC Gas Chromatography. 2 GC Mechanism of separation is primarily volatility. Difference in boiling point, vapor pressure etc. What controls.
ANALYTICAL CHEMISTRY CHEM 3811 CHAPTER 22 DR. AUGUSTINE OFORI AGYEMAN Assistant professor of chemistry Department of natural sciences Clayton state university.
Gas Chromatography Introduction 1.)Gas Chromatography Mobile phase (carrier gas) is a gas - Usually N 2, He, Ar and maybe H 2 - Mobile phase in liquid.
ENVE 202 Dr. Aslıhan Kerç GAS CHROMATOGRAPHY. Gas Chromatography (GC) *Gas chromatography is a chromatographic technique that can be used to separate.
Reading assignment: section 26E(p781) Chapter 26 # 2, 3, 14, 15, 16 Chapter 27 # 7(a,d,f), 22, 23, 24, 25.
CHAPTER 29 Supercritical Fluid Chromatography The mobile phase is a supercritical fluid (a fluid above its critical T and critical pressure) Supercritical.
Gas Chromatography. Gas Chromatography Basics Gas Liquid Chromatography (GLC) Gas Solid Chromatography (GSC) Mobile phase does not interact with analyte.
INSTRUMENTAL ANALYSIS CHEM 4811 CHAPTER 12 DR. AUGUSTINE OFORI AGYEMAN Assistant professor of chemistry Department of natural sciences Clayton state university.
LECTURE 4: CHROMATOGRAPHY Content: - Chromatographic separation - classifying analytical separations; column chromatography, planar chromatography - gas.
Gas Chromatography 427 PHC. Gas Chromatograph Gas chromatography (GC), is a common type of chromatography used in analytic chemistry for separating and.
Basic Gas Chromatography. History Separation of dyes by Runge Separation of plant pigments by Tswett Theoretical gc (Martin & Synge)
1/15 Gas Chromatography Haiqing Guo Dept. of Fire Protection Engineering Lab Methods Day June 25, 2014.
18-1 Gas Chromatography Vaporization of sample §Gas-solid àPhysical absorption §Gas-liquid àLiquid immobilized on inert solid Principles Instrumentation.
CHROMATOGRAPHY A laboratory technique in which the components of a sample are separated based on how they distribute between two chemical or physical.
GAS CHROMATOGRAPHY. Classification based on Mobile Phase Gas Chromatography Gas - solid Gas - liquid Stationary Phase Sample MUST be volatile at temperatures.
Experiments in Analytical Chemistry -Gas Chromatographic analysis of a gas mixture.
1 Gas Chromatography Lecture a. Thermal Conductivity Detector (TCD) This is a nondestructive detector which is used for the separation and collection.
Gas Chromatography Chap 27 Types: Gas-solid chromatography (GSC) Gas-solid chromatography (GSC) Limited application Limited application Gas-liquid chromatography.
Gas Chromatography. In Gas Chromatography (GC), a gaseous mobile phase transports a gaseous solute through a long, thin column containing solid or liquid.
Gas Chromatography A.) Introduction : Gas Chromatography (GC) is a chromatographic technique in which the mobile phase is a gas. GC is currently one of.
Chromatography vObjective To understand the principles of chromatography and know the specific types of Chromatograph used in the analysis of environmental.
Chapter 4. Theoretical Background Gas Chromatography HPLC Quantitation, Calibration, Standardisation and Validation.
Gas Chromatography A.) Introduction : Gas Chromatography (GC) - chromatographic technique where the mobile phase is a gas. GC is currently one of the most.
Gas Chromatography Carrier Gas Flow Control Injector Port Column Column Oven Detector Recorder The GC system consists of gas supplies for the mobile phase.
Gas chromatography is used in many research labs, industrial labs (quality control), forensic (arson and drug analysis, toxicology, etc.), environmental.
Gas Chromatography. Mobile phase: Inert gas such as N2 or He. Mobile phase: Inert gas such as N2 or He. Stationary phase: May be solid (GSC) or Stationary.
Ch 21 – Principles of Chromatography and Mass Spectrometry Ch 22 – Gas and Liquid Chromatography.
Analysis of Residues. Laboratory Analysis of Debris and Other Samples 1. Preparation of Liquid Samples: Liquid samples are simply drawn into a special.
The geometry of capillary columns is fairly simple, consisting of length, internal diameter, and stationary phase thickness. Nevertheless, there are endless.
Effect of Packing Particle Size on Plate Height. Resolution between 2 adjacent peaks.
High Performance Liquid Chromatography. The chromatogram is a record of detector output Vs time as the analyte passes through the chromatography.
Chapter 6 - Chromatography Week 3, Lesson 3. Chromatography Chromatography is a technique that is used to separate the substances present in a mixture.
CHROMATOGRAPHY. The general name given to methods by which two or more compounds in a mixture are physically separated by distributing themselves between.
Principles of Chromatography. Chromatography is the most powerful tool for separating & measuring the components of a complex mixture. Quantitative &
Chromatography Chapter Dr Gihan Gawish Principles Partition of molecules between liquid (stationary phase) and gas (mobile phase)
Chapter 22 GC & LC Gas Chromatography -1 1.Schematic diagram.
GAS LIQUID CHROMATOGRAPHY Principles Partition of molecules between gas (mobile phase) and liquid (stationary phase).
GAS CHROMATOGRAPHY In gas chromatography (GC), the sample is injected onto the head of a chromatographic column and immediately vaporized. The components.
In carbon-13 NMR, what do the number of peaks represent? The number of chemically different carbon atoms present.
Presented by – ANAMIKA BANERJEE (RESEARCH SCHOLAR)
Chapter 29 Supercritical Fluid Chromatography and Extraction.
Created with MindGenius Business 2005® Gas Chromatography Gas Chromatography From Harris, Quantitative Chemical Analysis, 6e, Chapter 24.
Gas Chromatography & Gas-Liquid Chromatography. Both gas chromatography and gas-liquid chromatography work in a very similar way.
Experiment 6 Simple and Fractional Distillation.
Standard Methods for the Examination of Water and Wastewater, 21 st Ed. (APHA AWWA 2005)
Intensive General Chemistry Chemical separations II Isabelle Vu Trieu
Chapter 32 Gas Chromatography. In gas chromatography, the components of a vaporized sample are separated by being distributed between a mobile gaseous.
© 2017 SlidePlayer.com Inc. All rights reserved.