3 Blood pump Moves blood in all the therapies. In CRRT the blood flow is between 0.30 and 450 ml/min.In CPFA and PEX the blood flow ranges from 0.30 to 250 ml/min.
4 Ultrafiltration PumpMoves the ultrafiltration fluid, ranging from 0.5 to 12 lt/h.Variable- Wanted weight lossReached weight loss
5 Plasma PumpIt moves Pre-dilution infusion, range Lt/h, in CVVH/ IHF and Post-dilution infusion, range Lt/h, in CVVHD/CVVHDF/IHD/IHDF.In CPFA the plasma flows in a range from 0.5 to 25 % of blood’s flow
6 Infusion PumpIn CVVH/ IHF/ CPFA - moves post-dilution infusion. In CVVHD/ CVVHDF/ IHD/ IHDF - moves dialysate. In PEX - moves substitution fluid.Flow ranges from 0-12 Lt/hAllows control over the precent of pre to post dilution infusion (green to blue pump flow).
7 Citrate pumpThis pump delivers citrate into the blood line.
8 7 Pressure Transducers7 pressure transducers check different pressures:arterial - from to +30 mmHgvenous - from -30 to +300 mmHgprefilter - from 0 to +400 mmHgtransmembrane (TMP) - from -300 to +400 mmHgplasma outputcartridge outputcartridge input
9 Blood Leak DetectorOptic detector with 2 sensors, can show a blood leak on the filtrate line.
10 Air detector and clampAn ultrasonic sensor reports the presence of air bubbles if bigger than 50 µl. This sensor is connected with a clamp to stop the venous blood flow in case of alarm.
11 Syringe PumpA pump for administration of continuous (0-10 ml/h) or bolus (0-20 ml) of anticoagulant or Ca2+.The pump can accept syringes of all types with a variable volume between 20 to 60 ml.
12 HCT and SO2 MeterThe Hematocrit (with blood volume calculation) and the Saturation are measured continuously through a high-precision instrument from the arterial line. There is also the possibility of checking the hemo-concentration through the direct measure of HCT.
13 Infusion Scale It controls the infusion-dialysate fluid. It can hold up to 40 liters exchange.
14 Ultrafiltration Scale It controls the ultrafiltrate fluid.It can hold up to 40 liters exchange.
15 HeaterIt allows to heat the infusion fluid in pre or post dilution. The temperature can be set between 30°C and 40°C with steps of 0.1°C.
16 Display touch screenA high-visibility display with user friendly and intuitive touch-screen interface. Allows clear, fast and direct dialogue with the machine. It shows colors and a clear graphic line with its 10.5 inches.
17 DisplayThe screen is divided into 4 areas in which information appear.Title area : name of active window and treatment.Function keys : they are used toactivate different functions.Active window :displays information and allows the user to set all data.On line guide
Your consent to our cookies if you continue to use this website.