4CHARACTERISTICSThe ACPR Reactor has two irradiation channels. The central tube ( Φ=304mm, h=228mm) and the pneumatic rabbit in D10 channel.In the stationary operation mode at 100KW in the central tube the thermal neutron flux is:Φth = 5*1011n/cm2*secand the cadmium ratio for this location is:Rcd = 1.15In the same operation conditions the unperturbed thermal neutron flux in the pneumatic rabbit channel is:Φth = 7.6*1011n/cm2*secand the cadmium ratio is:Rcd = 1.78
5The TRIGA SSR REACTOR 1 2 3 4 5 6 7 8 9 10 11 12 A B C D E F G H I J K L45L39L38L2H34R-7H3H28R-3L44R-2H21H26H22H30L32L46L47L49H25H12H14H18R-1L42L40R-4H17H6ABCDEFGHIJK
6CHARACTERISTICSThe maximum thermal neutron flux available in one of the in core location, namely XC-1 water filled, is:Φth = 2.7*1014n/cm2*secand the cadmium ratio is:Rcd = 3.08
7f = Φth/ Φepi = 42.3 ( measured using Zr foil detector) The thermal flux in the beryllium holes for 10MW reactor power.Location: a)J7: Φth = 6.4*1013n/cm2*secRcd = 2.3f = Φth/ Φepi = 15.2 (measured using Zr foildetector)b) K11: Φth = 9.54*1012n/cm2*secRcd = 4.15f = Φth/ Φepi = 42.3 ( measured using Zr foil detector)
8IRRADIATION DEVICESFor ACPR Reactor: A pneumatic rabbit for TRIGA ACPR reactor. The dimensions of cartridge used in this pneumatic rabbit are : Φ14 h190mm. The pneumatic rabbit may be operated manually or automatically timed.SSR Reactor: For the beryllium channels, is available another pneumatic rabbit with cartridge dimensions of : Φ12 h=30mm. This pneumatic rabbit is manually operated.Another irradiation device, also of pneumatic rabbit type, is provided with an underwater storage rack having a capacity of 12 cartridges. It also has an automatic timing control.
9GAMMA RAY SECTROMETERS AND SOFTWARE ANALYSIS PACKAGE Three high-resolution gamma ray spectrometers are available. All gamma ray detectors are HpGe type having relative efficiency of ~20% and V~100cmc.Two of the spectrometry chains are provided with an AQUSPEC multi-channel analyzer and GENIE2000 software analysis package.
10NAA METHODFor NAA the method of multi-isotope foil standardization ( Zr method) was implemented. The method was checked through SRM samples irradiation ( basalt rock, code BE-N) and of a sample kit supplied by Reactor Experiments. Same samples were also measured through ICPMS. The samples are weighted by the means of METTLER TOLEDO AX205 scale.The QA system is also implemented in the current activities of the laboratoryThe laboratory accreditation process is undergoing.