Presentation on theme: "APEX /IntegUSB Digital Front End Acquisition Interface and Nuclear Medicine Processing Software for APEX Gamma Cameras."— Presentation transcript:
APEX /IntegUSB Digital Front End Acquisition Interface and Nuclear Medicine Processing Software for APEX Gamma Cameras
Benefits Interfaces directly with Detector Signals (analog summer) More Stable –Analog Signal path is reduced to a minimum, no signal drifts. Less Power // Less Heat // Less Noise –Replaces all Analog Front End boards and Digital Correction & Processor Hardware, Less power consumption, No fans required. Smaller Footprint –Bulky Consoles can be removed. Better Specs –Final Camera Specs are usually better than the original (analog front end) camera.
IntegUSB DSP Based Front End Valid Event Detection Real Time High Speed Integration of all six channels Pulse Pile-Up detection and correction. Pre Window Energy Discrimination Position Normalization Event Buffering
IntegUSB Description Six Analog Input Channels. Six High Speed / High Resolution ADCs. Last generation DSP. High Count Rate (>500Kcps), Low Dead Time (1 µs Typ.) USB 2.0 (480 Mbps) interface to IM512P Acquisition Software. Analog ECG Interface for standard ECG Monitor / Gate –Digital input for R wave synchronization. –Analog input for ECG waveform display. Digital ECG Interface (For FiberOptic Isolated ECG-RMT)
Modifications to original Apex Camera Original APEX Gantry and Detector remain unmodified. We remove ALL Consoles, Monitors and Keyboard. We remove ALL Boards inside de Floor Cabinet. We add a last generation DSP based Front End and a DAC board to control PMTs and HV. We use existing HV and LV power supplies. We add a new opto-isolated ECG gating module. We add a last generation PC that hosts the IM512P Acquisition and Processing Software linked by a high speed USB2.0 interface. We provide a New Linearity Dot Mask, for Linearity Map acquisition. All Image Corrections are done in real time by our software.
Block Diagram – IntegUSB+IntDAC –> APEX IntegUSB A/D Valid Event Detection Pulse Pile-Up Corrector Integrator Position Normalization Buffering Digital & Analog ECG Interface X+ X- Y+ Y- E APEX Gamma Camera Detector 37, 59, 61 & 95 PMTs Heads Optical Transceiver Enorm IntDAC (mounted on top of IntegUSB) DACs to control PMT Gain and HV Auxiliary Gantry Control PMT Gain Control HV Power Supply HV Control USB 2.0 (480Mbps) PC with IM512P Acquisition & Processing Software (On the fly Energy, Linearity, Uniformity and C.O.R. Corrections) Digital ECG Module (battery powered) Versatile Link Optical Connection Patient Leads Digital ECG Input Gantry Gantry Control Ethernet Link to Optional IM512P Processing Workstations and/or DICOM Devices