Presentation on theme: "fMRI Quality Assurance (QA)(v2)"— Presentation transcript:
1fMRI Quality Assurance (QA)(v2) Data acquisition, processing and analysisKatherine LymerMedical Physicist in MR SupportSFC Brain Imaging Research Centre for Scotland, University of Edinburgh
2fMRI QA: Objectives Describe the quality of the fMRI data Using measures of SNR, SFNR, fluctuation and driftMonitor the long-term stability of scanner performance in fMRIAllow between site-assessments of fMRI data
3Image Acquisition Equipment set-up Test-object set-up Coil: As per usual fMRI examination (e.g. quadrature / 8 channel-head coil)Test object: Manufacturer provided homogenous sphereTest-object set-upMark as “zero” positionCentre coil / test object in scannerAlign lasers with centre of coil;Use test object holder (if applicable);Place test object in centre of coil;
4Image Acquisition Frequency of acquisition: Scanning parameters Site dependant: May be useful to acquire data on a weekly / fortnightly basis to establish a baseline; The results from our QA suggest that a monthly test is acceptable on our scanner.Scanning parametersNote magnet room temperatureSequence: Two approaches:As per typical fMRI examination“Pushing” the scanner as hard as possible (measure stability “under stress”)In Edinburgh, we use the following protocol on our 8 channel head coil:TETRNEXFOV#slicesSlice thicknessspacingnVolsMatrixFlip angleTscan40ms2044ms124cm285mm202*128*6490º6min 53sec* Includes dummy scans (processing software expects 2 dummy vols)Note the RF-receiver and transmit gain and the resonant frequency
5Image Processing Requirements: Transfer data to “off-line” to image analysis work stationNB This processing protocol is based upon that provided by the BIRN (http://www.nbirn.net/research/function/fbirn_tools_primer.shtm)and is similar to that used in “Calibrain”.See also:Weisskoff et al, MRM 36: , 1996Friedman et al, JMRI 23: 827–839, 2006Requirements:PC / workstation with matlab;SPM5 (as supplied by the FIL:fMRI_qa.m script (as supplied by fBIRN:This script needs to be modified to reflect local / in-house needs(4) mricro
6Data ManagementEnsure that data is stored in central fMRI data store, in appropriate folder denoting the current QA session(E.g. In Edinburgh, the data is stored in Y:\MONTHLY_QA using the format: QAddmmyyyyTEMPtt_yyyymmdd_studyNumber)In the relevant monthly QA folder, make two new sub-folders to differenentiate between the raw and reconstructed data e.g. “raw_data” and “SPM5_output”.Copy the fMRI dicom files into the raw_data folder(E.g. in Edinburgh, this is: QAddmmyyyyTEMPtt_yyyymmdd_studyNumber\raw_data)
7Pre-processing Start Matlab (ensure all paths are correct) Start SPM5 Convert the DICOM files using the “DICOM Import” function in SPM5:Select the files to be reconstructed (all in the “raw data” folder)Select “Done”;Select output dir: SPM5_output;Select “Done(NB if performing reconstruction on a windows machine:In pop-up window:Output image format: Two file (img +hdr) NIFTIUse ICEDims in filename: No)This may take a few minutes
8Pre-processing (4) Start mricro Check the central slice in volume Load the first “real” fMRI volume (i.e. the third listed volume) into mri croReview entire volume and note central sliceAmend fMRI_qa.m accordingly
9Image Processing: fMRI_qa_calc Edits required to fMRI_qa (original version obtained fromEnsure TR is correct;Ensure that the date is in the correct format;Define dimensions of ROI for ghosting measurement by changing:X1G, X2G, Y1G, Y2G(4) Ensure that the central ROI is placed appropriately within the central image(5) Check that the central slice number is correct (i.e. it matches that practically acquired)
10Image Processing: fMRI_qa_calc Run your version of fMRI_qaImport values into excel spreadsheet;Plot results to help identify any trends / outliers.Note any variation in RF-receiver / -transmit gain and resonant frequency (since the phantom and the protocol remain the same, these should remain constant)
11Image Analysis:All image analysis is performed on the central slice of the volume. (NB It is important to check that the central slice as designated in KL_fMRI_qa.m matches that practically acquired.)Application of the BIRN / Calibrain protocol yields six measurements:Mean signalA mean image is produced by calculating the mean, on a voxel by voxel basis, across all the central slices.A signal summary value is obtained from an ROI placed in the centre of this mean image.Example mean image:
12Image Analysis: (2) Temporal Fluctuation Noise Image A fluctuation noise image is produced by subtracting the calculated trend line from the data (the BIRN function uses a second order polynomial). (It’s effectively a standard deviation image.)By removing the trend from the data, we can estimate the fluctuations in the data about that trend: A linear trend may indicate a systematic increase or decrease in the data (caused by e.g. sensor drift).Example standard deviation image:
13Image Analysis:(3) Signal-to-Fluctuation Noise Ratio (SFNR) Image and Summary SFNR ValueA SFNR image is created by dividing the mean signal image by the temporal fluctuation (SD) image on a voxel-by-voxel basis.The summary SFNR value is obtained from the mean SFNR from the same ROI placed in the centre of the SFNR image.Example SFNR image:
14Image Analysis: (4) Static Spatial Noise In order to measure the spatial noise, the sum of the odd and even numbered slices are calculated separately (so there are two results); the difference between these two sums is then calculated. This approximates the static spatial noise.Is there is no drift in either amplitude or geometry across the time series, then this difference image will show no structure of the phantom and provides a measure of intrinsic noise.Example difference image:
15Image Analysis: (5) SNR Summary The same ROI is placed in the centre of the static spatial noise (difference) image.Using the signal summary value (see (1)), the
16Image Analysis: (6) SNR Percent Fluctuation and Drift The average signal intensity is calculated from each of the central slices across the time series using the same ROI, placed in the centre of each image.A time series of volume number vs average intensity is plotted and a second order polynomial is fitted to the data to establish a trend line.The mean signal intensity (prior to detrending) and the SD of the residuals after detrending are calculated.
17Image Analysis: (7) Summary graphs In addition, a series of summary graphs showing the change in raw signal, SGR, magnitude spectrum and relative standard deviation will be displayed
18Image Analysis: (8) Signal-to-Ghost Ratio (SGR) In addition to the BIRN tests, a signal to ghost (SGR) measurement was added to the protocol by Dr Gordon Waiter, Aberdeen Biomedical Imaging Centre, University of Aberdeen.The image may have to be windowed to observe the ghost.In order to make the SGR measurement, a second (ghosting) ROI is defined within the ghosting region of the image. The SGR summary value is calculated from the mean signal (from the central ROI) / mean signal (from the ghosting ROI).
19SINAPSE Reporting Mechanism: Short-termMaintain results locally;Distribute summary results with colleagues e.g. for multi-centre studies, comparisons of scanner performance (KL happy to hold central data base and distribute as required).Medium - Long termWe aim to produce a SINAPSE QA reporting website to which these results can be uploaded.Watch this space…………..!