Download presentation

Presentation is loading. Please wait.

Published byDeja Cripps Modified over 4 years ago

1
1 Chelmsford Amateur Radio Society Advanced Licence Course Carl Thomson G3PEM Slide Set 21: v1.0, 27-Sept-2004 (10) Measurements-1 - Meters Chelmsford Amateur Radio Society Advanced Course (10) Measurements Part-1 - Meters

2
2 Chelmsford Amateur Radio Society Advanced Licence Course Carl Thomson G3PEM Slide Set 21: v1.0, 27-Sept-2004 (10) Measurements-1 - Meters Types of Movement & Constructional Details Meter movements for DC and AC measurements can be moving-coil, moving-iron, hot-wire and thermocouple. The latter are good for measuring RF amps. For DC and low frequency AC, we will only consider the moving coil type. For AC a diode rectifier is needed in the meter circuit The maximum movement possible is called Full Scale Deflection, FSD SN

3
3 Chelmsford Amateur Radio Society Advanced Licence Course Carl Thomson G3PEM Slide Set 21: v1.0, 27-Sept-2004 (10) Measurements-1 - Meters Ammeter Example for 1 Amp FSD scale:- Calculate the voltage drop across the meter from V = I.R We get V = 1*10 -3 x 100 = 0.1 Volts. If 1mA flows through the meter then 0.999 amps must flow through shunt. There is 0.1 volts across the shunt. As R = V / I, then the shunt is 0.1 / 0.999 = 0.1001 Ohms. The basic meter movement is typically 1mA for FSD and has internal resistance of 100 ohms. Its range has to be extended to the required range by a Shunt Resistor, which be calculated by OHMS LAW. MeterInternal r Shunt R

4
4 Chelmsford Amateur Radio Society Advanced Licence Course Carl Thomson G3PEM Slide Set 21: v1.0, 27-Sept-2004 (10) Measurements-1 - Meters Voltmeter To measure voltages consider the basic meter movement with typical 1mA FSD and internal resistance of 100 Ohms. This is extended to the required voltage range by a series resistor. Called a Multiplier Resistor. The volts drop across the meter is 0.1 Volts so to have a full scale of 1V it is required to drop (1 - 0.1) Volts across the series resistor Apply OHMS LAW: R = V / I, with V = 1 - 0.1 = 0.9 Volts and I = 1 mA Therefore the multiplier resistor is V / I = 0.9 /.001 = 900 Ohms. - and the meter is a total of 1000 Ohms input resistance Meter Internal r Multiplier Resistor

5
5 Chelmsford Amateur Radio Society Advanced Licence Course Carl Thomson G3PEM Slide Set 21: v1.0, 27-Sept-2004 (10) Measurements-1 - Meters Ohmmeter To set for ZERO OHMS the two probe leads red and black are joined and the calibration R adjusted for FSD. Measuring a resistance introduces another series resistor into the circuit which reduces the current through the meter. Thus the scale can be calibrated in Ohms. Red Meter Internal r Calibration R + Black

6
6 Chelmsford Amateur Radio Society Advanced Licence Course Carl Thomson G3PEM Slide Set 21: v1.0, 27-Sept-2004 (10) Measurements-1 - Meters Meter Loading The lower 470k is shunted by 1000 Ohms so the bottom pair becomes; 470k x 1000 Ohms / (470 k + 1000) Ohms = 997.79 The volts measured will be 997.79 / 470997.79 x 1 = 0.002 volts. Oops! Care must be taking when measuring voltages across large resistors. Use a high impedance digital voltmeter or a very high ohms per volt analogue voltmeter. NB: DVMs may be >10MOhm impedance Potential Divider opposite should have a voltage at the midpoint of 0.5 Volts Total resistance of the meter is 1000 Ohms; 100 Ohms internal resistance plus 900 Ohms multiplier resistance. 470K 1V Internal Resistance & Multiplier

Similar presentations

Presentation is loading. Please wait....

OK

Measurement of Voltages and Currents

Measurement of Voltages and Currents

© 2018 SlidePlayer.com Inc.

All rights reserved.

To make this website work, we log user data and share it with processors. To use this website, you must agree to our Privacy Policy, including cookie policy.

Ads by Google