Presentation on theme: "Module 3 Use, calibration and maintenance of laboratory equipment"— Presentation transcript:
1Module 3 Use, calibration and maintenance of laboratory equipment
2Learning objectives At the end of this module you will be able to: describe briefly the operating characteristics of biological safety cabinets, centrifuges, autoclaves, inspissators, pH meters, incubators;correctly use biological safety cabinets, centrifuges, autoclaves, inspissators, pH meters, incubators;describe the main maintenance operations for biological safety cabinets, centrifuges, autoclaves, inspissators, pH meters, incubators.
3Content outline Principles, use and maintenance of: biological safety cabinet (BSC)centrifugeautoclaveinspissator (coagulator)pH meterincubator
4Biological safety cabinet (BSC) A ventilated contained area providingprotection for the operator and theenvironment against infectious aerosolsduring the handling of hazardous microorganismsThe most important equipment in all diagnostic mycobacteriology laboratories is the BSC.
5HEPA filter HEPA = high-efficiency particulate air The HEPA filter traps and remove 99.97% of airborne particles of diameter 0.3 µm or more.
6BSC class I and II – main differences BSCs should be ducted or vented to the outside.No recirculation into the room is allowed.The BSC should be connected to a UPS (Universal Power Supply).BSC class I and II – main differencesBSC classMinimal facevelocity(m/s)Air flow (%)Re ExhaustedcirculatedExhaust systemI0.38100Hard ductIIA17030Thimble connectionIIA20.51IIB1IIB2
7Class I BSC Protection of the user and the environment Air is ducted to the outside of the building.HEPA filter
8Class II BSC Protection of the user, the environment and the product Class IIA BSCClass IIB BSC
9Laminar flow hood for product (not user) protection Horizontal airflowLaminar vertical flowAir blows towards workers.Used only to protect product for sterile preparations (media preparation).
10BSC locationA , B – not recommendedD, C – recommendedDCAB
11BSC – operations for correct use Switch on the BSC airflow and the UV light (if present) 5–15 minutes before use.Check the airflow velocity on the display.Complete the logbook, recording details of operation of the BSC.Switch off the UV lamp.Never work with the UV lamp on.Disinfect the work surface, interior walls and interior window surface with 70% ethanol or other suitable compounds.Do not use corrosive substances.
12BSC – rules for correct use Organize all items needed for work in the BSC to avoid frequent displacements.Organize the work to minimize arm movements.Do not overcrowd the working area – this disturbs the airflow.Conduct all manipulations within the BSC as far towards the back of the unit as possible (at least 15cm from the front grille).In class II BSCs, never allow the front grille to be covered with anything.Do not use large open flames in the BSC.Do not accumulate waste in the BSC; remove it when activities are finished.
13BSC – after use Autoclave the waste. Wipe down the inside of the BSC and work surface with 70% alcohol.Switch on the UV light for 1 hour (if the lamp is in good working order).Switch off the BSC fan.
14Certification and maintenance Regular certification and maintenance, with replacement of filters, are needed and represent a major challenge in low-income countries. If maintenance is not carried out regularly, the filters may become clogged and tubercle bacilli may be blown into the face of the operator. A BSC that is not well maintained is more hazardous than protective.
15BSC – evaluation of performance Performance of the BSC should be evaluated by the manufacturer or a qualified professional, with specific, well-calibrated equipment.
16BSC – maintenanceModern cabinets are equipped with airflow indicators and warning devices.Maintenance, with replacement of filters, should be carried out by professionals when the airflow falls below the minimum recommended level.The BSC should not be used until this maintenance has been carried out.Out of order
17BSC – maintenanceAirflow smoke pattern tests are performed to determine:whether the airflow along the entire perimeter of the work access opening is inward;whether airflow within the work area is downward with no dead spots or refluxing;whether ambient air passes onto or over the work surface;whether there is refluxing to the outside at the window wiper gasket and side seals.The smoke test is an indicator of airflow direction, not of velocity.Commercial airflow testers are recommended. They are small glass tubes, sealed at each end. Both ends are broken off with the gadget provided and a rubber bulb fitted to one end. Pressing the bulb to pass air through the tube causes it to emit white smoke.
18BSC – maintenance A cabinet leak test is carried out: to determine whether exterior surfaces of all plenums, welds, gaskets, and plenum penetrations or seals are free of leaks;before initial installation or annually on fully installed cabinets
19BSC – fumigation Decontamination of the BSC with formaldehyde gas. Hydrogen peroxide.Fumigation should be performed only by professionals.System DECONTAKIT
20BSC – fumigation Fumigation of the BSC should be carried out: before release of the BSC for use after a major biohazardous spill;before replacement of HEPA filters;before repair work requiring access to the sealed plenum;before service or replacement of the circulation fan or components;before maintenance work in contaminated areas;before performance tests requiring entry into contaminated areas;before movement of the BSC to another laboratory or area of use;before changing work programmes, e.g. to non-TB work;before release of the BSC for resale or salvage.
23Centrifuge – selection of proper equipment for TB culture The centrifuge should:Have aerosol-free (O-ring sealed) swing buckets.Have locking mechanism for protection to prevent opening before rotation has ceased.Be able to generate 3000g.Be refrigerated.
27Centrifuge – useInspect the centrifuge chamber for liquid or signs of corrosion.Check that there are shock-absorbing pads in the bottom of the centrifuge buckets.Balance the sample tubes and insert them in the buckets in the appropriate position.Never add water to a specimen to balance paired tubesUse an empty tube filled with water as a “balance” .Select the required time and speed.Stop centrifuge immediately if any abnormal noise is noticed.Never operate a centrifuge with the lid open.Open the sealed buckets in the BSC.
28Centrifuge tubes – adaptors Centrifuge tubes must tolerate g-forces of at least 3000g.Centrifuge tubes must be used with appropriate rubber or plastic cushions matched to the tube and bucket-holder.Use only adaptors recommended by the manufacturer of the centrifuge in use.
29RCF, g, RPM RCF (or g) = relative centrifugal force rpm = revolutions per minuteRCF = 1.12 * 10−6 * r * v2orRCF (g) = 1118 x r x (v/1000)2r = radius in mmv = speed in rpmrpm = 1000 *√RCF/1.12 * rAlways work in RCF, not rpm: newcentrifuges can convert automatically.r (mm)
30Centrifuge – installation Ensure that the centrifuge is installed on a rigid, flat, level surface.Because centrifuges produce vibration during operation, they must not be installed next to balances or other sensitive equipment.Allow sufficient free space around the centrifuge for ventilation to prevent overheating.Initial calibration should be performed only by a qualified service technician .
31Centrifuge – daily maintenance Wipe the inside bowl with disinfectant solution and rinse thoroughly.If a refrigerated centrifuge is turned off at night, open the top to allow the bowl to dry. During the day when the unit is under refrigeration, keep the top closed to avoid condensation and ice build-up.
32Centrifuge – monthly maintenance Clean the centrifuge housing, rotor chamber, rotors and rotor accessories with a neutral cleaning agent.Clean plastic and non-metal parts with a fresh solution of 5% sodium hypochlorite (bleach).
33Centrifuge – annual maintenance Annual service should be carried out by a qualified service technician: centrifuge brushes, timer, speed and electrical leaks should all be checked.
36AutoclaveThe autoclave using saturated steam under pressure is the most efficient means of:sterilization in a diagnostic TB laboratory;decontamination of biological material consisting of infectious waste.
37Autoclave For optimum function: All of the air in the chamber should be replaced by steam.The temperature must be 121 ºC.Materials to be sterilized must be packed loosely.
38Tips for loading the autoclave Only appropriate containers should be used.Discarded cultures should be in solid-bottomed containers no more than 40 cm deep.Leave a large air space around each container.Never cover containers.If autoclaving closed containers, add water to generate steam.Separate autoclaves should be used for sterilization of solutions or glassware (clean materials) and for decontamination of infectious materials.
39Autoclaves suitable for laboratory use Pressure-cooker laboratory autoclaveAutoclave with air discharge by gravity displacement
41Autoclave – usePrepare material for autoclaving with thermosensitive paper.Do not load material to be sterilized with material to be decontaminated.Fill the bottom of the autoclave with distilled water.Open the air outlet valve.Turn on the heating.Close the outlet valve.Do not touch the drainage tap or the outlet or the safety valve while heating under pressure.When the required time is up, turn off the heating completely.When the temperature falls below 100 ºC, open the outlet valve slowly.
42Autoclave – useNever unscrew the lid clamps and open the lid until the hissing sound has stopped.Leave the sterilized/decontaminated material to cool before removing it from the autoclave.Check whether the autoclave tape has turned black and the covering paper has turned brown. If they have not, the material is not decontaminated. Check the autoclave for malfunction.Biological indicators should be used after 40 hours of use.
43Autoclave – suggested protocols Solid material for sterilization: 121 ºC (appropriate pressure 115 kPa), 20 min.Solid material for decontamination: 121 ºC (appropriate pressure 115 kPa), 30 min.Liquid material for sterilization: 15 min at 121 ºC if “liquid cycle” is available.
44Autoclave – maintenance Check door gaskets for cracks and pitting.Check for proper operation of door clamps and door locks.Inspect valve discs and seats for signs of wear or cutting.Check pressure-release safety valves and thermometers.
47InspissatorEssential when egg media are “home-made” (made in the laboratory).
48Inspissator – use Carefully control the amount of heating. Ensure the required temperature has been reached before loading the tubesHeat the tubes at constant temperature of 80–85 ºC for 45 minutes.Tubes or bottles loaded at an angle of 5– 10º ( ° ).Always check the quality of media after cooling.
49Inspissator – maintenance Check temperature at each use.Clean after preparation of each batch of culture media.
53pH meter – useCalibrate the pH meter before use at three pH values (pH 4.0, 7.0 and 10.0), using the calibration solutions provided by the manufacturer.Immerse the electrode in the test solution.Rinse the electrode after use.Cover the pH meter after use.
54pH meter – calibrationThe pH meter must be calibrated according to the manufacturer’s instructions.Calibration must be performed once daily or, in case of infrequent use, at least on the day of pH testing. It must be performed before the first measurement of the day.Standardize with pH 4.0 and 7.0 buffers before each test or series of tests.Discard contaminated or cloudy standard buffers.Calibration results are acceptable if the pH values of the reference buffer solutions are within 0.1 pH units of the expected pH.
55pH meter – MaintenanceKeep the instrument clean. Cover it after each use.Rinse the electrode after use.Make sure that the electrode is always filled with electrolyte according to the manufacturer’s instructions.Do not touch the electrode membrane, which can be easily damaged.A glass electrode should be kept immersed in a standard salt solution for long-term storage.Calomel electrodes must be kept in a potassium chloride (KCl) buffer solution when not in use.Date buffer solutions and discard when unsatisfactory.
56pH-meter – calibration and record sheet Calibration solutionSolution testDateTemp.(ºC)Electrodeserial no.pH 4pH 7pH8*SignatureSolution/bufferpH read*pH 10 can be used if commercially purchased
57pH-meter – logbook ITEM IDENTIFICATION FAILURE EVENTS Equipment Brand namePurchase dateModel/typeLocation within the labSerial no.Warranty expiration dateManufacturerAddressTel.Contact personTechnical service representativeFAILURE EVENTSDateEventCorrective action takenOperator
58IncubatorsIncubators ensure optimum growth conditions for culture of M. tuberculosis.Incubators are available in various sizes, from small (benchtop) models to large closets up to several hundred liters. Walk-in incubator rooms with circulating fans to obtain an homogeneous temperature are also frequently used.
59Incubator – useKeep door(s) closed to prevent heat loss to the environment.Ensure that rack positions are clearly marked.Switch the incubator off when not in use.Do not overload.
60Incubator – maintenance Record temperatures daily on appropriate forms.Check that the actual temperature corresponds with the thermostat setting.Clean every 14 days with 70% ethanol.Clean immediately after any infectious spills.
61Incubator – calibration Calibrate:before use;after temperature changes have been detected and rectified;following a power failure;after cleaning of spillages.
62Incubator – calibration Ensure that the door is closed and that the incubator is switched on.Set the required temperature using the temperature control and leave the incubator to run for 1 hour.Place a thermometer in the centre of the incubator with the probe away from the heating element.Take the temperature reading after 30 minutes. If the temperature is not 36 ± 1 ºC, adjust the control.Repeat this process every 30 minutes until the required temperature is reached.Continue to take readings until two consecutive readings (30 minutes apart) are 36 ± 1 ºC.
65True and false exercise The biological safety cabinet is the most important equipment for TB culture laboratories.Only class II BSCs can be used for TB culture.Centrifuge speed should always be expressed in rpm.Centrifugation may produce hazardous aerosols.Checking the autoclave regularly is unnecessary if it is used only for decontamination.Equipment maintenance is crucial for optimal performance.Maintenance can always be performed in-house.
66Module review: take-home messages Equipment is expensive: all equipment must be treated with care and properly maintained.Quality-control checks on equipment should be run regularly and results should be recorded on appropriate forms.The biological safety cabinet is the most important item of equipment for TB culture under safe conditions.Read carefully and always follow manufacturers’ instructions for routine work and maintenance.Familiarize yourself with all working parts of all the instruments.Call for help when troubleshooting any equipment malfunction.
67Self-assessment Describe the differences between the two types of BSC. What precautions must you take when working in a BSC?Describe the operations to be undertaken before starting a centrifuge.Explain the principles of sterilization using an autoclave.