Download presentation

1
**Calibration Techniques**

1. Calibration Curve Method 2. Standard Additions Method 3. Internal Standard Method

2
**Calibration Curve Method**

Most convenient when a large number of similar samples are to be analyzed. Most common technique. Facilitates calculation of Figures of Merit.

3
**Calibration Curve Procedure**

Prepare a series of standard solutions (analyte solutions with known concentrations). Plot [analyte] vs. Analytical Signal. Use signal for unknown to find [analyte].

4
**Example: Pb in Blood by GFAAS**

Signal (ppb) (mAbs) 0.50 3.76 1.50 9.16 2.50 15.03 3.50 20.42 4.50 25.33 5.50 31.87 Results of linear regression: S = mC + b m = 5.56 mAbs/ppb b = 0.93 mAbs

6
**C = (27.5 mAbs – 0.92 mAbs) / 5.56 mAbs / ppb (3 significant figures)**

A sample containing an unknown amount of Pb gives a signal of 27.5 mAbs. Calculate the Pb concentration. S = mC + b C = (S - b) / m C = (27.5 mAbs – 0.92 mAbs) / 5.56 mAbs / ppb C = 4.78 ppb (3 significant figures)

7
**Calculate the LOD for Pb**

20 blank measurements gives an average signal 0.92 mAbs with a standard deviation of σbl = 0.36 mAbs LOD = 3 σbl/m = 3 x 0.36 mAbs / 5.56 mAbs/ppb LOD = 0.2 ppb (1 significant figure)

8
**(include this point on the calibration curve)**

Find the LDR for Pb Lower end = LOD = 0.2 ppb (include this point on the calibration curve) SLOD = 5.56 x = 2.0 mAbs (0.2 ppb , 2.0 mAbs)

9
**This is approximately 5% below the expected value of 103.71 mAbs**

Find the LDR for Pb Upper end = collect points beyond the linear region and estimate the 95% point. Suppose a standard containing 18.5 ppb gives rise to s signal of mAbs This is approximately 5% below the expected value of mAbs (18.50 ppb , mAbs)

10
**Find the LDR for Pb LDR = 0.2 ppb to 18.50 ppb or**

LDR = log(18.5) – log(0.2) = 1.97 2.0 orders of magnitude 2.0 decades

11
**Calculate the slope of the log-log plot**

Find the Linearity Calculate the slope of the log-log plot

12
Not Linear??

13
Not Linear??

14
**log(S) = log(m) + log(C)**

Remember S = mC + b log(S) = log (mC + b) b must be ZERO!! log(S) = log(m) + log(C) The original curve did not pass through the origin. We must subtract the blank signal from each point.

15
Corrected Data

16
Linear!

17
**Standard Addition Method**

Most convenient when a small number of samples are to be analyzed. Useful when the analyte is present in a complicated matrix and no ideal blank is available.

18
**Standard Addition Procedure**

Add one or more increments of a standard solution to sample aliquots of the same size. Each mixture is then diluted to the same volume. Prepare a plot of Analytical Signal versus: volume of standard solution added, or concentration of analyte added.

19
**Standard Addition Procedure**

The x-intercept of the standard addition plot corresponds to the amount of analyte that must have been present in the sample (after accounting for dilution). The standard addition method assumes: the curve is linear over the concentration range the y-intercept of a calibration curve would be 0

20
**Example: Fe in Drinking Water**

The concentration of the Fe standard solution is 11.1 ppm All solutions are diluted to a final volume of 50 mL

21
-6.08 mL

22
**Vsam x [Fe]sam = Vstd x [Fe]std**

x-intercept = mL Therefore, 10 mL of sample diluted to 50 mL would give a signal equivalent to 6.08 mL of standard diluted to 50 mL. Vsam x [Fe]sam = Vstd x [Fe]std 10.0 mL x [Fe] = 6.08 mL x 11.1 ppm [Fe] = 6.75 ppm

23
**Internal Standard Method**

Most convenient when variations in analytical sample size, position, or matrix limit the precision of a technique. May correct for certain types of noise.

24
**Internal Standard Procedure**

Prepare a set of standard solutions for analyte (A) as with the calibration curve method, but add a constant amount of a second species (B) to each solution. Prepare a plot of SA/SB versus [A].

25
Notes The resulting measurement will be independent of sample size and position. Species A & B must not produce signals that interfere with each other. Usually they are separated by wavelength or time.

26
**Example: Pb by ICP Emission**

Each Pb solution contains 100 ppm Cu.

28
**No Internal Standard Correction**

29
**Internal Standard Correction**

30
**Results for an unknown sample after adding 100 ppm Cu**

Similar presentations

OK

CHEM 213 Instrumental Analysis Lab Lecture – Copper by AA & Least Squares Analysis.

CHEM 213 Instrumental Analysis Lab Lecture – Copper by AA & Least Squares Analysis.

© 2018 SlidePlayer.com Inc.

All rights reserved.

Ads by Google

Free download ppt on alternative sources of energy Ppt on ip address classes Ppt on computer languages 2015 Ppt on unity in diversity in india Ppt on ip address classes subnet Ppt on semi solid casting Ppt on applied operational research society Ppt on history of islam Ppt on pin diode limiters Ppt on unity in diversity india