Presentation on theme: "Spiro Vision 3+ This slide show will advance automatically. To manually advance, click on the screen, or press enter."— Presentation transcript:
1Spiro Vision 3+This slide show will advance automatically. To manually advance, click on the screen, or press enter.
2If you want to stop the presentation and go back, here are the instructions...... To go back to a slide, press either:Left Arrow ( ) or the Up Arrow ( )
3Hooking up the handsetInsert turbine into handset and gently push it in all the way.
4Plug the SpiroVision-3+ cable into the computer’s USB port
5To verify that the software settings are correct: Select OptionsSelect Settings
6Make sure the following are selected: Interface: USBDevice Selection: Spirovision USBUnit of meas.: Inch/LbsPress “OK” to exit.If you have changed a setting, close and open SpiroVision-3+ again.
7Adding A New PatientIn the main screen, select the “Archive Navigator” button from the tool bar.
8Adding A New PatientThe Archive Navigator Screen is where patient files are stored. Click on the “Add A New Patient” icon to fill in the Patient Card.
9Entering Patient Information Enter the patient ID number, name, date of birth, race and sex in the Patient Card and click OK. Other fields are optional and can be customized.
10In the next screen, enter the patient’s height and weight In the next screen, enter the patient’s height and weight. To display the patient’s “lung age”, enter smoking information. Click on OK to continue.Most predicted norms are calculated based on the patient’s age, race, sex and height.
11Review the patient information in the Archive Navigator and click on OK.
12Performing an FVC TestTo start testing, click on the “Execute Test” icon.Select “Forced Vital Capacity Pre”A new window will pop up with a graphThe software will verify that the handset is connected and display a little green circle on the lower right.
13Place a filter or mouthpiece on the SpiroVision-3+ handset.
14Start by instructing the patient on how to perform an FVC maneuver: Seal your lips around the mouthpiece or filter.Take 2-3 normal breaths.Inhale deeply and completely fill your lungs.Blast out as hard as you can. Keep exhaling until you can’t any more – try to go at least 6 seconds!(for a full loop, continue) Inhale as fully and with as much force as possible.
15The Bubbles IncentiveTo display the Bubbles Pediatric Incentive on screen select “View” , then “Encouragement”.
16When the patient is ready, start the test by clicking on the START button
17As the patient exhales, you will see the flow/volume and time/volume graphs in real time.
18As the patient exhales, you will see the flow/volume and time/volume graphs in real time.
19As the patient exhales, you will see the flow/volume and time/volume graphs in real time.
20Printing ResultsOnce you have saved the results the graphs and data will be displayed on the screen
21As the patient exhales, you will see the flow/volume and time/volume graphs in real time.
22Once the test is finished the Quality Control window will let you know how the maneuver compared to ATS standards for spirometry.You can “accept” or “discard” each test individually; there will be future options for discarding an effort.
23Following the maneuver, the curve and actual values of the test can be compared with the predicteds.
24Saving TestsTo repeat the test, select START. The maneuver can be repeated as many times as desired. A minimum of 3 efforts is recommended.At the end of the testing session select EXIT.
25All the tests performed during the session will be listed (from best to worst). Highlight the one (or ones) you’d like to save in the patient record and click OK.
26Printing ResultsTo print, select the “Print” icon, or press “Ctrl P”.
27Reports can be generated through your printer or saved as a pdf file. They can be customized and are in color.Reports meet ATS, OSHA, NIOSH, and SSD standards for spirometry.
28Thank you for taking the time to learn more about SpiroVision-3+ Thank you for taking the time to learn more about SpiroVision-3+. If you have any questions please call us at