Presentation on theme: "Support Part II EMROL – Your power partner Support – Part two What are profiles and an explanation of the parameters How to improve the accuracy of a."— Presentation transcript:
Support – Part two What are profiles and an explanation of the parameters How to improve the accuracy of a capacity test Charger and cycle test on a battery Use of monitor test The possibility of recording data in a system with a load/source on a battery Can the AlfaBat break down, how can you test it (tools and technical manual) and how can you fix it (spare parts + instructions)
EMROL – Your power partner Profiles What are profiles? What are parameters? 3 profiles in the AlfaBat, Unlimited on the PC How to use them? Always use the same profile or choose to select one Upload to device
EMROL – Your power partner Improving the accuracy Correct battery temperature Battery quality Peukert value
EMROL – Your power partner Peukert Peukert capacity = capacity when discharged with one ampere Normal value = 1.10 to 1.36 Ideal value = 1 High values indicate poor battery quality AlfaBat has a function to determine this factor
EMROL – Your power partner Charger and cycle test on a battery Select a correct profile Adaptor is required Connect the charger Charger test: result in Ah Usefull to record a chargers regime. Cycle discharge: result in % Cycle charge: Charge factor
EMROL – Your power partner Charge Factor Relation between the discharged capacity and the charged capacity > 1.10 : to compensate charge losses you need to put more energy in a battery then what was extracted.
EMROL – Your power partner Monitor Test Online tracking of a test in progress Necessary during cycle test (unless printer is used) Needed for peukert calculation Alternative to cycle test with PC: use of the printer
EMROL – Your power partner Recording data in a system Connect the source/load to the charger port. Make sure the load respects the max current 15A or 50A. Connect the battery to the battery port Start a recording test (use the correct profile)
EMROL – Your power partner Technical solutions 1. Simple things first: all connections are present, polarity ok, bad connection, latest firmware / software,... 2. Look in the knowledge base on the Emrol website. 3. If you are sure there is a technical problem look in the technical manual 4. Use the service tools to locate / fix the technical problem
EMROL – Your power partner Technical Manual Use ONLY original spare parts (FAN, Fuse, controller board,...). Use one of the tools for testing the device in combination with the Windows software.
EMROL – Your power partner AlfaBat Service Programming
EMROL – Your power partner AlfaBat Service Power supply
EMROL – Your power partner AlfaBat Service Measuring
EMROL – Your power partner AlfaBat Service Test Peripheral
EMROL – Your power partner AlfaBat Service Device