2THE SUPERCHARGED SKC AIRCHEK XR5000 The FIRST personalsampling pump withsupercharged lithium-ionbattery power.Experienceunparalleledrun times up to 40 hours on onebattery charge when runningat 2 L/min with thehigh-power Li-Ionbattery pack and 0.8 µmMCE filter.SKC has the FIRST and ONLY sampling pump that uses powerful lithium-ion batteries. That is what separates us from all the rest. We capitalize this in our marketing by using the phrase:Li-Ion battery makes it flow, like no other sampling pump.Most of you report that Gilian is your main pump competitor. So the Gilian 5000 would be the direct competitor to the XR I included the Gilian spec sheet in your handout. This pump uses nickel metal hydride and they are quoting standard run times of “8hours minimum”.The lithium-ion battery would be a good thing to get written to specs to lock out the competition.
3SKC AIRCHEK XR5000 Designed for what customers really need: Xtended Range-up to 5 L/minXtended Run Times-up to 40 hours at2 L/min with high-power Li-Ion battery; 20 hours with standard Li-Ion battery.Xtra Easy Operation-3-button controlAfter doing some customer surveys, SKC determined whatcustomers really wanted in a pump was:Simple operationPowerThis pump is NOT a replacement for the Airchek TheAirchek 2000 pump will continue to be the pump of choice forhigh tech users who want PC compatibility.I see the XR5000 as the pump of choice for routine, dailyhygiene, indoor air, or fenceline sampling.
4AIRCHEK XR5000 SPOTLIGHT ON FLOW Patented internal flow sensorAutomatically compensates for changes in air temperatureConstant flow in high flow modeMulti-tube sampling in low flow mode using low flow adapter kitThe XR5000 uses the patented isothermal flow sensor to maintain constant flow. But unlike the AC2000, the XR5000 does not display the flowrate. Instead there is just a blue LED, that blinks to indicate the pump is running.We made this decision to keep the operation simple and the cost down. We have already received feedback from some of you and other distributors, that you want an indicator of flow---either a rotameter or digital display on future pump designs. I agree. We have advised Product Development.The pump’s flowrate will compensate for changes in air temperature, but there is NO internal atm pressure sensor.Like other designs, uses CPC to do multi-tube sampling in low flow mode.
5AIRCHEK XR5000 SPOTLIGHT ON PROGRAMMING Timed Run-Manually program the desired sample time up to 9999 minutes with auto shut-off for ease of use in the field.Delayed Start-Begin sampling automatically during late-night shifts or other applications without the need for user start-up.Our market research told us that for routine day-to-day use that customers want the features that we offer on the PCXR4 pump. So the XR5000 features:Timed runDelay startLCD also has a battery status icon to indicate approximate battery charge remaining. This seems to be a very important feature. The only thing positive we hear about the TSI pump is that it tells you how much approx. run time is remaining on the battery.
6AIRCHEK XR5000 SPOTLIGHT ON SAMPLE INTEGRITY Lockable Keypad-eliminates the possibility of pump tamperingFault Mode-maintains sample time upon low battery or flow restrictionAuto-restart-attempts restart after 15 seconds in flow faultTo enhance sample integrity, the XR5000 has several features:LOCK OUT feature on keypadFAULT SHUTDOWN WITH TIME RETENTIONAUTO RESTART
7AIRCHEK XR5000 SPOTLIGHT ON SIZE Compact-standard model is only 8 x 6 x 11 cmLightweight-standard model weighs only 454 grams (16 oz).FIRST IMPRESSIONS are very important in everything!So I suggest to you that you use the XR5000 with standard battery as your demo pumps. They are very, very small and make a very good first impression.(Show side-by-side with Gilian 5000)
8AIRCHEK XR5000 SPOTLIGHT ON SAFEGUARDS CE marked for RFI/EMI shieldingAnti-static plastic caseParticulate trap on inletUL approved for intrinsic safetyRoHS compliantIn this last slide, the safeguards built into the XR5000 pump are listed.And now, I would like to welcome and thank the person MOST responsible for the last and most important bullet on this slide--UL approval of this pump:Dr. Saulius TrakumasSaulius will now give you an update on the XR5000 and other issues related to SKC sampling pumps.