Presentation on theme: "Fresenius Medical Care Peritoneal Dialysis Connections"— Presentation transcript:
1Fresenius Medical Care Peritoneal Dialysis Connections Safe-Lock to Baxter Connection
2Connections using Fresenius Medical Care products This inservice is not intended to replace the advice or training from your physician. Refer to the Instructions For Use (IFU) for detailed device description, instructions, contraindications, warnings, and precautions.Fresenius Medical Care, DEL-CLAMP, Stay●Safe, Safe-Lock, DELFLEX and FREEDOM are trademarks of Fresenius Medical Care Holdings Inc. or its affiliated companies. All other trademarks are the property of their respective owners.
3Fresenius ProductsBaxter is a trademark of Baxter International Inc. or its affiliates.Instructions on how to use Baxter products were reproduced from the manufacturer’s literature and were current at the time of print. Instructions are included for the patient's convenience but Fresenius Medical Care North America does not specifically endorse the use of these products or assume any liability related to their use. Should there be any differences between the instructions listed in this manual and the instructions for use provided with the medical devices referenced, the instructions provided with the medical device should be taken as accurate."
4Connections using Fresenius Medical Care products Note: for clarity the model is shown not wearing personal protective equipment (PPE). Follow your facility’s policies and procedures for use of personal protective equipment.It is vital to maintain sterility of the connectors during connection procedures.
5Safe-Lock patient to Baxter supplies A Universal Connector will allow the patient with a Fresenius Safe-Lock PD system to use Baxter supplies while in the hospital.Universal ConnectorFresenius Safe-Lock PD systemBaxter Supplies
6Safe-Lock SystemsPlease note Safe-Lock systems include:Delflex Premier Plus exchange setFreedom Y-set with snap disconnectAll Freedom cycler tubing setsNewton cycler set with safe lock connectors8 Lead Safe-Lock Manifold set
7Safe-Lock® patient to Baxter supplies Type II adapter on catheterUniversal ConnectorP/N #4 inch convertible extension setExtension set with roller clampPre-filledUniversal ConnectorP/N #Premier transfer setA Universal Connector (P/N # ) or Pre-filled Universal Connector (P/N # ) will connect to any of the patient catheter or catheter extension sets shown above.
8Safe-Lock patient to Baxter supplies Gather suppliesPre-filled Universal ConnectorBaxter tubing solution set and related supplies for transfer set connection and exchange procedure.Baxter extension set
9Safe-Lock patient to Baxter supplies 12Hold Premier transfer setUntwist tubing segment secured with DEL-CLAMP clamp.Remove tubing segment from patient’s catheter3
10Safe-Lock patient to Baxter supplies 44. Remove sealing capfrom large end ofUniversal Connector.5. Attach UniversalConnector to patient’s catheter6. Remove the sterility protectorfrom the small end of the56
11Safe-Lock patient to Baxter supplies 77. Remove the sterility protectorfrom the Baxter extension set.8. Attach the Baxter extensionset to the UniversalConnector. Proceed withexchange.8
12Safe-Lock patient to Baxter supplies Typical connection and use of the Universal Connector
13DISCHARGE INSTRUCTIONS FOR ALL CONNECTIONS The patient should return to his/her home or clinic with the additional extension in place. This practice will allow the patient to return home and resume dialysis without interruption.Instruct the patient to remove any adaptors added to their extension set at the beginning of the next exchange or as directed by the PD clinic nurse.
14Select the System the Hospital Uses Choose the type of catheter or transfer set and the type of system utilized. Look down the column and across the row to the adapter needed. Click on the icon in the box to take you to the procedureSelect the catheter or transfer set used by the patient.Select the System the Hospital UsesCatheterorTransfer Setor orWith Luer-Lock AdapterTitaniumWith Safe-Lock AdapterWith Baxter Transfer SetWith Fresenius Safe-Lock transfer SetWith Fresenius Stay-Safe Transfer SetHospital Uses Baxter SuppliesAddBaxter Transfer Set5C4484Add FMC Universal Connectoror pre-filledNo Change NeededAdd FMC Convertible Adapter and Baxter 5C4484Stay●Safe Luer Lock Adapter 4”Hospital Uses Fresenius (FMC) Safe-Lock SuppliesAdd Safe-Lock Transfer Set with Luer Lock AdapterNo Change Needed ORAdd a FMC Safe-Lock Transfer Set with a Safe-Lock AdapterAdd Convertible Extension4” Stay●Safe Safe-Lock AdapterHospital Uses FMC Stay-Safe SuppliesStay● Safe Extension Set with Luer lock Patient ConnectorStay● Safe Extension Set with Safe lock Patient ConnectorAdd Stay● Safe Luer Lock Adapter and a Baxter Transfer Set 5C4484Add Stay● Safe Extension set with Safe-Lock Patient connector
15Please contact your Fresenius Clinical Sales Representative at if you have questions regarding the use of a Fresenius PD system or adapter.