Presentation on theme: "Pulse Voltammetric Techniques"— Presentation transcript:
1Pulse Voltammetric Techniques The important parameters for pulse techniques are as follows:1. Pulse amplitude is the height of the potential pulse.This may or may not be constant depending upon the technique.2. Pulse width is the duration of the potential pulse.3. Sample period is the time at the end of the pulse during which the current is measured.4. For some pulse techniques, the pulse period or drop time must also be specified. This parameter defines the time required for one potential cycle, and is particularly significant for polarography
2Frederick Gardner Cottrell Born in Jan Frederick Gardner Cottrell Born in Jan. 10, 1877, Oakland, California, U.S.A. Dead in November 16, 1948, Berkeley, Calififornia, U.S.A.Frederick Gardner Cottrell was U.S. educator, scientist, and inventor of the electrostatic precipitator, a device that removes suspended particles from streams of gases. However, he is the best known for electrochemists because of the "Cottrell equation".
10Concentration Profile Fig. 2. Qualitative diagrams showing concentration--distance profile atvarious stage of the cyclic voltammogram shown in left Fig..
11Cyclic Voltammetry – ox + e ⇄ Red 1. the peak potential separation (Epa - Epc) is equal to 57/n mV for all scan rates where n is the number of electron equivalents transferred during the redox process.2. the peak width is equal to 28.5/n mV for all scan rates.3. the peak current ratio (ipa/ipc) is equal to 1 for all scan rates.4. the peak current function increases linearly as a function of the square root of v.
13Animated Cyclic Voltammetry Experiment Fc Fc+ + e-E(t) = Ei + v tNernst equation; E = Eo' + RT/nF ln aO /aR
14Example of Nernstian (reversible) Behavior Electrode area = 0.1 cm2ks = 1 cm/s, Eo = 0.3 VDO = DR = 1 x 10-5 cm2/sip = 2.69 x 105 n3/2 A DO1/2 v1/2 CO
15Determination of the Peak Current ip = 2.69 x 105 n3/2 A DO1/2 v1/2 COwhere:A is the electrode area (cm2 ),n is the number of electrons transferred,CO is the concentration (mol.cm-3 ),v is the scan rate (volt/s.)
17The Nernst Equation1. The Nernst equation is named after the German physical chemist Walther Nernst who first formulated it.2. The Nernst equation links the actual reversible potential of an electrode (measured in volts), E, to the standard reversible potential of the electrode couple, E0 which is a thermodynamic value.3. In its most fundamental form the Nernst equation is written as:R: the universal gas constant,T: the absolute temperature in degrees Kelvin,z : the charge number of the electrode reaction (which is the number of moles of electrons involved in the reaction as written),F: is the Faraday constant (96,500 C mole-1).The notation ared represents the chemical activities of all of the species which appear on the reduced side of the electrode reaction and the notation aox represents the chemical activities of all of the species which appear on the oxidized side of the electrode reaction.
18IntroductionThe charging current decays exponentially, whereas the faradaic current (for a diffusion-controlled current) decays as a function of 1/(time)½; that is, the rate of decay of the charging current is considerably faster than the decay of the faradaic current.The charging current is negligible at a time of 5RuCdl after the potential step (RuCdl is the time constant for the electrochemical cell, and ranges from µs to ms).Therefore, after this time, the measured current consists solely of the faradaic current; that is, measuring the current at the end of a potential pulse allows discrimination between the faradaic and charging currents
19Cyclic Voltammetry – ox + e ⇄ Red 1. the peak potential separation (Epa - Epc) is equal to 57/n mV for all scan rates where n is the number of electron equivalents transferred during the redox process.2. the peak width is equal to 28.5/n mV for all scan rates.3. the peak current ratio (ipa/ipc) is equal to 1 for all scan rates.4. the peak current function increases linearly as a function of the square root of v.
20Cyclic Voltammetryipc and ipa refer to the peak current of the cathodic and anodic sweeps.Epa and Epc refer to the potential (voltage) of the anodic and cathodic current peaks.
21A CV of a fully reversible system will display the following characteristics: The voltage separation between the current peaks isThe positions of peak voltage do not alter as a function of voltage scan rateThe ratio of the peak currents is equal to oneThe peak currents are proportional to the square root of the scan rate
22The influence of the voltage scan rate on the current for a reversible electron transfer can be seen below:
23The effect of differences in the kinetic rate (kred) constant for electron transfer on the LSV of a system are illustrated below: (Left)Both Epc and ipc shift to different values as electron transfer becomes more difficult. In cyclic voltammetry, a system can have different cathodic (Kred) and anodic (Kox) kinetic rate constant. Such systems are known as quasi-irreversible. The CV is illustrated below: (Right)
24Different pulse techniques A number of different pulse techniques are available on the epsilon, which differ in their potential pulse wave forms, the number of sampling points, and whether a solid electrode (voltammetry) or a mercury drop electrode (polarography) is used.The discrimination against the charging current that is inherent in these techniques leads to lower detection limits (when compared to linear sweep techniques), which makes these techniques suitable for quantitative analysis.Sampled Current PolarographyNormal Pulse Voltammetry/Polarography (NPV)Differential Pulse Voltammetry/Polarography (DPV)Square Wave Voltammetry (SWV)
25Potential wave form for sampled current polarography Potential is varied in a series of steps, with the current sampled at the end of each step
26Change Parameters dialog box for sampled current polarography Range of allowed parameter values:Potential = mV Step E = mV Quiet Time = sStep Width = ms (Polarography); ms (Voltammetry)
27A typical sampled current polarogram The limiting current (id) is given by the Ilkovic equation:id = 708 n C D1/2 m2/3 t1/6where:n = number of electronstransferred/moleculeC = concentration (mol cm-3)D = diffusion coefficient(cm2 s-1)m = mercury flow rate (mg s-1)t = sampling interval (s)
28Potential wave form for normal pulse voltammetry (NPV) ssssThe potential wave form consists of a series of pulses of increasing amplitude, with the potential returning to the initial value after each pulse
29Change Parameters dialog box for normal pulse voltammetry (NPV)
30Range of allowed parameter values: Potential = -3000 - +3000 mV Step E = mVPulse Width = msStep Width = ms (Polarography);ms (Voltammetry)Quiet Time = s
32Potential wave form for differential pulse voltammetry (DPV) ss(t)The potential wave form consists of small pulses (of constant amplitude) superimposed upon a staircase wave form. Unlike NPV, the current is sampled twice in each Pulse Period (once before the pulse, and at the end of the pulse), and the difference between these two current values is recorded and displayed
33Change Parameters dialog box for differential pulse voltammetry (DPV)
34A typical differential pulse voltammogram (DPV) Potential = mV ; Step E = mV; Quiet Time = sPulse Amplitude = mV.; Pulse Width = msStep Width = ms (Polarography); ms (Voltammetry)
35Simultaneous Determination of Dopamine and Ascorbic Acid at an in-site Functionalized Self-Assembled Monolayer on Gold Electrode L. Zhang, J. Jia, X. Zou, S. Dong*, Electroanalysis, 16(2004)1-61. The in-site functionalization of 4-aminothiophenol (4-ATP) self-assembled monolayer on gold electrode at physiological pH yields a redox active monolayer of 4’-mercapto-N-phenylquinone diimine (MNPD). Th functionalized electrode exhibits excellent electrocatalytic responses towards dopamine (DA) and ascorbic acid (AA), reducing the overpotentials by about 0.22 V and 0.34 V, respectively, with greatly enhanced current responses.2. Due to its different catalytic activities toward DA and AA, the modified electrode resolves the overlapping voltammetric responses of DA and AA into two well-defined voltammetric peaks by differential pulse voltammetry (DPV), which can be used for the simultaneous determination of these species in a mixture.3. The catalytic peak current obtained from DPV was linearly related to DA and AA concentration in the ranges of 5.0x x10-4 M and 8.0 x x10-4 M with correlation coefficient of and 0.998, respectively. The detective limits (3s) for DA and AA were found to be 1.2x10-6 M and 2.4 x10-6 M, respectively.
36The in-site functionalization of the 4-ATP SAM yields a redox active monolayer of 4’-mercapto-N-phenylquinone diimine (MNPD)
37Fig. 4. Cyclic voltammograms of 1. 0x104 M DA (A) and 1 Fig. 4. Cyclic voltammograms of 1.0x104 M DA (A) and 1.0x 104 M AA (B) at clear Au (dashed line) and MNPD/Au (solid line) in 0.1 M PBS (pH 7.2). Scan rate: 50 mV s1.Fig. 5. Differential pulse voltammograms of 1.25x104 M DA and 1.30 x104 MAA mixture at clear Au (A) and MNPD/Au(B) in 0.1 M PBS (pH 7.2).
38Fig. 6. A) DPVs of DA (40 mM) at MNPD/Au in 0. 1 M PBS (pH 7 Fig. 6. A) DPVs of DA (40 mM) at MNPD/Au in 0.1 M PBS (pH 7.2) in the different concentrations of AA (a ± f: 60, 75, 90, 105, 120, 135 mM). B) DPVs of AA (120 mM) at MNPD/Au in 0.1 M PBS (pH 7.2) in the different concentrations of DA (a ± g: 5, 20, 35, 50, 65, 80, 95 mM).Fig. 7. Differential pulse voltammograms of DA and AA mixtures at MNPD/Au in 0.1 M PBS (pH 7.2). DA contents from a) to g) are 5, 25, 45, 65, 85, 105, 125 mM, respectively. AA contents from a) to g) are 8, 25, 50, 70, 90, 110, 130 mM, respectively.
39Square Wave Voltammetry (SWV) The potential wave form consists of a square wave of constant amplitude superimposed on a staircase wave form. The current is measured at the end of each half-cycle, and the current measured on the reverse half-cycle (ir) is subtracted from the current measured on the forward half-cycle (if). This difference current (if - ir) is displayed as a function of the applied potential.
40Potential wave form for square wave voltammetry (SWV)
41Change Parameters dialog box for square wave voltammetry (SWV) Range of allowed parameter values: Quiet Time = sPotential = mV; Step E = mVS.W. Amplitude = mV; S.W. Frequency = Hz
44Experiment for SWV or DPV 5 mM ferrocyanide and 10 mM ferrocyanide in 0.1 M KCl1 mm glassy carbon electrode, Ag/AgClInitial Potential-100 mVFinal Potential650 mVPulse Amplitude50 mVStep Height2 mVPeriod20 msConcentrationPeak HeightPeak Area10 mM46.3 µA34.8 µC5 mM23.7 µA17.2 µC
45Picomolar Peroxide Detection Using a Chemically Activated Redox Mediator and Square Wave Voltammetry Jennifer L. Lyon and Keith J. Stevenson* Anal. Chem. 2006, 78,1. A method for low-level, low-potential electrochemical detection of H2O2 using a chemically activated redox mediator is presented. This method is unique in that it utilizes a mediator, Amplex Red, which is only redox-active when chemically oxidized by H2O2 in the presence of the enzyme horseradish peroxidase (HRP).Scheme 1. Depiction of Enzymatically Generated H2O2 Detection Using the Amplex Red Redox Mediatora
46Abstract2. Microelectrode square wave voltammetry was used to optimize sensing at ultralow concentrations (<1 mM), this method exhibits marked improvements in analytical sensitivity and detection limits (limit of detection as low as 8 pM) over existing protocols.3. Sensing schemes incorporating both freely diffusing and immobilized HRP are evaluated, and the resulting analytical sensitivities are 1.22±0.04 and (2.1± 0.6) 10-1 mA/(mMmm2), respectively, for peroxide concentrations in the high picomolar to low micromolar range. A second linear region exists for lower peroxide concentrations. Furthermore, quantitative enzyme kinetics analysis using Michaelis-Menten parameters is possible through interpretation of data collected in this scheme. Km values for soluble and immobilized HRP were 84±13 and 504±19 mM, respectively.4. This method is amenable to any biological detection scheme that generates hydrogen peroxide as a reactive product.
47Several drawbacks when using Pt as a working electrode for H2O2 oxidation Poor selectivity, low sensitivity and high susceptibility to electrode foulingThe electrochemical oxidation of generated H2O2 at Pt at physiological pH values ( ) occurs at potentials (+0.4 to +0.7 V vs Ag/AgCl) where other electroactive species (ascorbic or uric acid) typically found in biological samples interfereBecause the electrochemical oxidation of H2O2 produces protons, a pH gradient is generated at the surface of the Pt electrode, resulting in perturbation of biological samples that are sensitive to changes in pHQuantitative measurements become complicated as the oxidation of H2O2 at physiological pH is mechanistically complex, involving the formation of several reactive intermediates (superoxide, hydroperoxide, and hydroxyl radicals) whose stability is also pH dependent
48Amplex RedReactions of Amplex Red (N-acetyl-3,7-dihydroxyphenoxazine) and of resorufin (7-hydroxy-3H-phenoxazin-3-one).Electroanalysis, 17(2005)1043Fig. 2. Electrochemistry of Amplex Red and resorufin in pH 5 McIlvaine buffer. Cyclic voltam- mogram: starting potential þ0.1 V (vs. Ag/AgCl), scan rate 0.01 V s1, arrow follows the initial course of CV of Amplex Red, before the occurence of electrochemically generated resorufin.
49Electrochemical Performance of Amplex Red Fig. 4. H2O2 assay with 0.2 U mL1 HRP and 50 mmol L-1 Amplex Red performed in McIlvaine buffer pH 5 with CV detection. Only the first half-cycle of each CV is plotted: þ0.1 V to 0.7 V (vs. Ag/AgCl), scan rate 0.01 V s-1.Fig. 5. Amperometric flow-through detection of resorufin. Concentration dependence of peak current; glassy carbon electrode (diam. 3 mm) at 0.17 V (vs. Ag/AgCl), buffer flow rate 0.2 mL min-1, sample volume 0.1 mL, buffer deoxygenated.Electroanalysis, 17(2005)1043
50Comparison of the Behaviors of SWV with CV Fig. 1. (a) Comparison of square wave (heavy trace) and cyclic(light trace) voltammograms for a 300 nM H2O2 standard, demonstrating improved S/N attainable with SWV. SWV parameters: amplitude 25 mV, step height 5 mV, frequency 25 Hz. CV parameter: scan rate 125 mV/s. Experimental conditions: 10 mm diameter GC working electrode, Ag/AgCl reference electrode, Au wire counter electrode, 0.1 M K2HPO4/0.05 M citric acid buffer, pH 5.0, 0.2 U/mL HRP, 10 mM Amplex Red. (b) Comparison of square wave voltammograms for 300 pM H2O2 collected using the same electrode as in (a) (heavy trace) and a 2 mm diameter GC disk electrode (light trace). All other experimental conditions are the same as those for (a).
51Square Wave Voltammetry The dimensionless kinetic parameter K =ks/f is related to DEp values between 10 and 40 mVOverall rate constant of ks = ( 4.78 s-1 was determined for the resorufin-dihydroresorufin couple. This value is in accord with other reports for strongly adsorbing redox- active species studied via SWV.Fig. 2. Peak height vs. frequency data for a 1 nM H2O2 standard detected via Amplex Red-mediated SWV (R2 ) 0.994, SD=0.028 nA). Experimental conditions are the same as those listed for Fig. 1.
52H2O2 Calibration CurvesFig. 3. Calibration curves for H2O2 standards detected via Amplex Red-mediated electrochemical sensing, utilizing either soluble (open squares) or immobilized (closed squares) HRP. Peak currents were obtained using SWV experiments outlined in the text and normalized to the working electrode area. SWV parameters: amplitude 25 mV, step height 5 mV, frequency 25 Hz. Experimental conditions: 10 mM diameter GC working electrode, Ag/AgCl reference electrode, Au wire counter electrode, 0.1 M K2HPO4/0.05 M citric acid buffer, pH 5.0, 10 mM Amplex Red. For soluble HRP, [HRP] ) 0.2 U/mL. For immobilized HRP, an equivalent amount of HRP was adsorbed to a glass slide placed in the electrochemical cell. (a) Linear current response in the picomolar H2O2 range for soluble HRP. (b) Linear current response in the pico- to nanomolar range for immobilized HRP. (c) Wide-range linear responses for both soluble and immobilized HRP observed between 300 pM and low micromolar H2O2 concentrations. Sensitivities of these calibration curves are 3 orders of magnitude less than those in (a) and (b).
53Quantitative Enzyme Kinetics Measurements Fig. 4. Enzymatic activity data for Amplex Red-mediated electrochemical sensing. SWV parameters and experimental conditions are the same as those for Figure 3, except that the Amplex Red concentration was increased to 1 mM. (a) Full dynamic range for 15 pMto 1 mM H2O2, reflective of Michaelis-Menten enzymatic activity. (b, c) Lineweaver-Burke plots of current response data shown in Figure 3b for (b) soluble and (c) immobilized HRP. Statistics for (b): R2 ) 0.986, SD ) nA-1. Statistics for (c): R2 ) 0.996, SD ) nA-1.